- Research article
- Open Access
On-spot quantification of modafinil in generic medicines purchased from the Internet using handheld Fourier transform-infrared, near-infrared and Raman spectroscopy
Journal of Analytical Science and Technology volume 11, Article number: 35 (2020)
Poor quality medicines represent an expanding global public health threat facilitated by the Internet. A recent survey showed that one in five students have used modafinil to enhance learning ability mainly purchased from Internet sources. The aim of this work was to develop on-the-spot and simple methods for the quantification of modafinil in generic medicines using Fourier transform-infrared (FTIR), near-infrared (NIR) and Raman spectroscopy along with partial least square regression (PLSR). Modafinil tablets were measured in intact form using NIR and Raman and in powdered form using FTIR, NIR and Raman. Additionally, powder mixtures of crushed modafinil tablets and excipient(s) were prepared either by diluting the crushed tablets with excipient(s), or sequentially adding excipient(s) to the crushed tablets. Three PLSR models were constructed in MATLAB 2014a from powder mixtures and two from intact and powdered tablets. For FTIR and Raman spectroscopy, PLSR models based on tablets gave linear calibration curve with correlation coefficient (r2) values above 0.94 and a root mean square error of calibration (RMSEC) below 0.96% m/m. Conversely, the PLSR model based on powder sequential addition gave the highest accuracy using the NIR spectra (r2 = 0.99, RMSEC = 1.15% m/m). The latter model showed accuracy in predicting the concentration of the active pharmaceutical ingredient in modafinil generic medicines proving their authenticity. The overall results showed that the combination of the three spectroscopic methods with PLSR offered a rapid technique for authenticating generic modafinil medicines.
Poor quality medicines represent a global threat to the public health that can result in treatment ineffectiveness, drug resistance, increased morbidity and mortality rate, economic loss and problems to the healthcare system (Gaurvika et al. 2019; Janvier et al. 2018). Poor quality medicines can be degraded, substandard or counterfeit medicines (Mukhopadhyay 2007). Degraded medicines include those, which deteriorate from the poor-quality storage (humidity, temperature and light). Substandard medicines are those that encounter accidental defects in the manufacturing process and fail to fulfil the products’ specifications. Counterfeit medicines are medicines which “are deliberately and fraudulently mislabelled with respect to identity and/or source” (WHO Medicines 2012).
The Internet plays a major role in the spread of poor quality medicines, which could be over the counter products, prescription medicines, drugs of abuse and supplementary products (FDA The possible dangers of buying medicines over the Internet 2014; Fittler et al. 2013; Orizio et al. 2011). This is partly due to the fact that the market of counterfeit online medicines is in continuous expansion (Orizio et al. 2009). According to the World Health Organisation (WHO), 50% of the medicines sold through illegal online pharmacies are counterfeits (WHO Counterfeit medicines 2006). Thus, in 2013, the Interpol closed down over 9600 illegal online pharmacies and seized over 9.6 million prescription medicines that were worth over $41.1 million (Safemedicines 2013). In this respect, the purchase of a counterfeit medicine could impose a public safety issue especially in case of drugs of abuse where medicines are frequently bought online.
Smart drugs represent one of the top classes that sales have increased (up to 50%) over the last decade and has been facilitated by the Internet (International Narcotics Control Board 2016). Smart Drugs such include nootropics that have become particularly popular among students and healthcare professionals who have been under pressure of study/work (Champagne et al. 2019; International Narcotics Control Board 2016). Students and healthcare professionals have utilised nootropics for enhancement of the memory and learning abilities (Baskys and Remington 1996; Turner et al. 2003; Gold and Balster 1996). There are several different nootropics on the market, e.g. adderall, dexedrine, ephedrine, methylphenidate, modafinil and piracetam. Modafinil has the same stimulant effects of nootropics; however, it has less abusive tendencies and side effects (Baskys and Remington 1996; Turner et al. 2003).
Modafinil is sold under the brand name Provigil (Young-Powell and Page 2014) and comprises modafinil as the active pharmaceutical ingredient (API) and the following excipients: croscarmellose sodium, lactose monohydrate, magnesium stearate, maize starch and povidone. Nonetheless, the excipients in generic modafinil are not always known, and this variation influences the process of authentication of branded and generic modafinil. Thus, when authenticating a branded medicine, the test medicine needs to match the physical and chemical properties of the reference medicine (The Medicines Compendium Modafinil Provigil 100 mg Tablets 2020). Generic medicines however only need to prove that they contain the exact API at the correct concentration in relation to the reference medicine (Assi et al. 2012). Consequently, a quantitative approach in authenticating generic modafinil medicines is favoured.
The literature reported conventional methods for quantification of the API in modafinil, which range from simple spectrophotometric to chromatographic methods. A spectrophotometric method was reported for the quantification of modafinil in solid dosage forms and underlined measurement of the absorbance of modafinil at its maximum wavelength (236 nm) (Moffat et al. 2009). Chromatographic methods utilised mainly reverse phase high performance liquid chromatography (RP-HPLC) (Moffat et al. 2009; Venkatesh et al. 2011; Nageswara Rao et al. 2007; Cass et al. 2001), and thin layer chromatography (TLC) (Xu et al. 2002). The aforementioned techniques offered sensitivity, selectivity and precision yet they were time-consuming, destructive and required extensive sample preparation. On the contrary, spectroscopic techniques including Fourier transform-infrared (FTIR), near-infrared (NIR) and Raman spectroscopy have shown to be quicker, simpler and mobile (Faisall et al. 2009; Assi et al. 2011a; Assi et al. 2011b; Caporaso et al. 2018; Bouyé et al. 2018; Correia et al. 2018; Yang et al. 2019; Fedchak 2014; Crocombe 2018; Gerace et al. 2019). When combined with multivariate regression analysis, spectroscopic techniques offered rapid, on-spot and non-destructive quantification of APIs medicines (The Medicines Compendium Modafinil Provigil 100 mg Tablets 2020; Crocombe 2018). To the best of our knowledge, no spectroscopic methods have yet been employed for quantification of modafinil in tablets.
Therefore, this work aimed at developing methods for the on-spot quantification of modafinil in generic medicines using FTIR, NIR and Raman spectroscopy along with partial least square regression (PLSR).
Materials and methods
Standard reference material including glucose, lactose, magnesium stearate, maize starch, microcrystalline cellulose, modafinil, povidone, sodium carboxymethylcellulose and sucrose were purchased from Sigma-Aldrich.
Eight modafinil generic batches of doses 100 and 200 mg were bought from four Internet websites (Table 1). The percentage mass per mass (% m/m) of modafinil in the tablets was in the range of 57.6–72.7% m/m.
Reference analysis of modafinil API and tablets was performed using reverse phase-high performance liquid chromatography (RP-HPLC) by adopting the procedure given by Rao et al. 2007 (Rao et al. 2007).
Four types of samples were considered in this study. The first type included intact tablets, which were removed from the packaging and used ‘as received’ without any treatment. The second type comprised powdered tablets which had been crushed in a mortar, homogenised and stored in 4 mm glass vials. The third type of samples comprised powders of pure substances (API and excipients) and the fourth type included powdered mixtures that were prepared by mixing crushed modafinil tablets with excipient(s).
Three modafinil mixtures were prepared and included: M1 (modafinil lactose dilution), M2 (modafinil excipients dilution) and M3 (modafinil excipients sequential addition) (Table 2). M1 (modafinil lactose dilution) was prepared by adding aliquots of lactose (major excipient in modafinil tablets) to crushed modafinil tablets to get a % m/m of modafinil in the range of 9.59–62.5% m/m. Similarly, M2 was prepared by adding aliquots of different excipients (one at a time) to crushed modafinil tablets to get 15.4–52.9% m/m of modafinil. A third approach was adopted in mixtures (M3) which was made by adding excipients (one at a time) sequentially to aliquots of crushed modafinil tablets to get 15.9–62.5% m/m.
FTIR spectra were recorded using the Bruker Alpha mobile-FTIR equipped with a single reflection pure diamond attenuated total reflectance (ATR) crystal sample interface. The spectral range of the aforementioned instrument was 500–6000 cm−1. NIR spectra were recorded employing the JDSU microNIR 1700 pro-spectrometer equipped with linear variable filter (LVF) dispersing element and 128-pixel cooled InGaAs photodiode array detector. The NIR spectra were measured over the wavelength range of 900–1700 nm. Raman spectra were recorded using the Rigaku FirstGuard handheld Raman spectrometer equipped with 1064 nm laser power, thermoelectric cooling and charge coupled device detector. Raman spectra were collected over the wavenumber range of 250–2000 cm−1.
For FTIR measurements, a few milligrams from powdered samples or pure substances were measured by placing them in direct contact with the ATR crystal. Homogeneous preparations of samples were prepared using a Vortex mixer before each measurement. Four spectra were measured per sample such that a new aliquot was changed after each measurement. Each spectrum was the sum of 16 scans at a resolution of 4 cm−1. For NIR and Raman measurements, intact tablets were measured ‘as received’ by placing them in direct contact with the spectrometers. Four spectra were taken from each tablet on both sides, such as two spectra were taken from each side rotating the tablet after each measurement. In addition, powders were measured through glass vials (after mixing with Vortex mixer) by placing them in direct contact with the instruments. Each spectrum was the sum of 32 scans for NIR and 3 scans for Raman, respectively.
Spectra from the three instruments were exported to MATLAB 2014a for analysis. Spectral pre-treatment was made using multiplicative scatter correction second derivative (MSC-D1). The similarity between spectra was assessed using correlation in wavelength space (CWS) method. In this respect, a correlation coefficient (r) value greater than or equal to 0.95 showed similarity. In addition, quantitative models were developed using PLSR. PLSR has been considered as ideal in this case where univariate regression had not been possible. This was because the absorbance and scattering intensities in FTIR/NIR and Raman differed according to the physicochemical properties of the measured samples and not proportional to the concentration of the analyte of interest (Burns and Ciurczak 2007). In this respect, PLSR offered a multivariate approach for quantifying the APIs in the aforementioned products. PLSR models predicted the concentrations of the different mixtures and/or products from multiple variables (absorbance intensities or scattering intensities measured at the full wavelength range). PLSR models find components (latent variables) in the absorbance and/or scattering intensities that relate to the concentrations. A PLSR model eventually assigns loadings (small and large) to the aforementioned latent variables. Latent variables with small loadings are rejected and vice versa. This is made by finding factors that capture variation among the data such that each factor is added as one at a time. In this sense, the first factor captures the highest variance, the second factor the second highest variance and so on. The following equations illustrate a PLSR model (Brereton 2003; Jee 2019):
X is the absorbance or scattering intensities at different wavelengths
c is the concentrations
q is the loading vector
T is the spectral score vector
p is the spectral loading vector
Results and discussion
The present study explored a swift quantification of medicines purchased from several Internet sources using handheld instruments. This was the first study that had utilised quantitative PLSR models (non-destructive) with portable handheld FTIR, NIR and Raman spectroscopy, as well as a powder form of formulations with FTIR for the quantification of modafinil in branded and generic tablets. The aforementioned PLSR models were not only limited to conventional dilution models but also included more complex mixtures based on standard and sequential additions of constituents to crushed tablets.
Eight modafinil products (from different batches) were purchased from four websites. The eight products were selected based on assessing the differences of authenticity of products between websites as well as within each website. Four of these products had a label claim of modafinil 100 mg; while the remaining four had a label claim of modafinil 200 mg. The concentration range of modafinil in the four products was 57.6–72.7% m/m (Table 1). The tablets were compared in relation to the major constituents (API and excipients) expected to be present in branded and generic modafinil tablets. Excipients present in branded modafinil tablets (Provigil) include lactose monohydrate (main excipient), pregelatinised maize starch, croscarmellose sodium, povidone K29/32 and magnesium stearate (Gold and Balster 1996). The excipient content of generic tablets may be variable and not always known (Xu et al. 2002); therefore, additional excipients were measured including glucose, maize starch, microcrystalline cellulose, sodium carboxymethylcellulose and sucrose. The spectra of modafinil tablets were compared to the spectra of the API, excipient(s) and caffeine using the three techniques.
FTIR, NIR and Raman activity of modafinil tablets
Prior to spectral evaluation, the FTIR, NIR and Raman activity of modafinil tablets and their main constituents had been investigated. When comparing the three spectroscopic techniques in relation to medicines’ identification, it is well known that APIs are more Raman active whereas excipients are more IR/NIR active where the Raman activity of excipients is often masked by fluorescence (Faisall et al. 2009). If the medicine contains high concentrations of an excipient then the Raman activity of the medicine could be masked by the fluorescence exhibited by the excipient. One way of overcoming fluorescence of excipients was by using a longer wavelength laser such as 1064 nm, and that had been adopted in the current work.
The FTIR, NIR and Raman spectra of the medicinal products were compared to those of the API (modafinil) and the major excipient (lactose monohydrate) in modafinil tablets. Modafinil API was present in high amounts in all of the measured products (57–72% m/m) which minimised the effect of the excipients (Faisall et al. 2009). This was confirmed in the spectra of modafinil products, its API and lactose using the three techniques (Fig. 1). In this respect, the modafinil tablets’ spectra showed representation of the modafinil API rather than lactose. The FTIR spectrum of modafinil tablet (Fig. 1a) showed higher similarity for the API spectrum (r = 0.95) than the lactose spectrum (r = 0.82). Likewise the modafinil tablet NIR spectrum showed higher representation for the API spectrum (r = 0.99) than lactose spectrum (r = 0.77). The modafinil Raman spectrum showed higher similarity for the API spectrum (r = 0.95) but dissimilarity to the lactose spectrum (r = 0.01). This could be attributed to the strong Raman activity of modafinil API that had not been affected by the fluorescence of lactose. Subsequently, the high representation of the API in the tablets’ spectra contributed to the accuracy of quantification of tablets. Crushing the tablets into powders was needed to facilitate data collection, and while this process may affect the physical properties of the powder (such as the particle size), our observations showed that the spectroscopic data were not affected in this particular case. Some properties such as polymorphic nature of API are likely to be affected if strong physical processing was applied; however, in our experiments, we had used gentle processing to ensure minimal energy had been applied to the tablets. Such delicate processing avoided any polymorphic changes (such as recrystallisation or amorphous formation). Hence reproducibility was not affected by sample preparation.
PLSR model construction
PLSR was applied to the MSC-D1 FTIR, NIR and Raman spectra over the full wavenumber/wavelength in each technique. Four models were created using the FTIR spectra and five models were created using the NIR and Raman spectra (Table 3).
FTIR models included: FTIRM1 (modafinil lactose dilution), FTIRM2 (modafinil excipients dilution), FTIRM3 (modafinil excipients sequential addition) and FTIRM4 (modafinil powdered tablets model). FTIRM1, FTIRM2 and FTIRM3 were constructed using a calibration validation (C:V) ratio of 2:1. Moreover, the calibration ranges used were 9.49–62.5, 15.4–52.9 and 15.9–62.5% m/m, respectively. The modafinil powdered tablet model (FTIRM4) was constructed with a C:V ratio of 3:1, four factors and a range of 54.9–62.4% m/m.
NIR models included NIRM1 (modafinil lactose dilution), NIRM2 (modafinil excipients dilution), NIRM3 (modafinil excipients sequential addition), NIRM4 (modafinil powdered tablets model) and NIRM5 (modafinil intact tablet model). NIRM1, NIRM2 and NIRM3 were constructed with a C:V ratio of 2:1 and had calibration ranges of 9.49–62.5, 15.4–52.9 and 15.9–62.5% m/m, respectively. In addition, NIRM4 and NIRM5 were created with a C:V ratio of 3:1 and calibration range of 54.9–62.5% m/m, respectively.
The Raman models used were RamanM1 (modafinil lactose dilution), RamanM2 (modafinil excipients dilution), RamanM3 (modafinil excipients sequential addition), RamanM4 (modafinil powdered tablets model) and RamanM5 (modafinil intact tablet model). RamanM1, RamanM2 and RamanM3 were made with a C:V ratio of 2:1 and had calibration range of 9.49–62.5, 15.4–52.9 and 15.9–62.5% m/m, respectively. Furthermore, RamanM4 and RamanM5 were constructed with a C:V ratio of 3:1 and calibration range of 54.9–62.5% m/m, respectively.
PLSR model validation
The linearity of the models was evaluated by internal validation criteria calculated using the calibration and internal validation sets. For internal validation, the criteria considered were the regression correlation coefficient (r2), root mean square error of calibration (RMSEC) and root mean square error of prediction (RMSEP) of the internal validation set. The r2 and RMSEC were calculated by interpreting the relationship between the predicted concentration and the nominal concentration of the calibration set. Likewise, the RMSEP was calculated by interpreting the relationship between the predicted concentration and the nominal concentration of the validation set. If the model was a good fit, the relationship would be linear and an r2 value close to 1 would be obtained. There was no optimal value for the RMSEC and RMSEP; however, the lower they were, the more accurate was the model. A more accurate judgement was made by evaluating the relative standard error of prediction (RSEP), which was calculated as the percentage of the ratio of RMSEP to the mean value of the prediction set. A threshold value of ± 5% was taken for RSEP.
For FTIR models, the highest accuracy was observed for FTIRM1 (modafinil powdered tablet model), which showed r2 values of 0.98 and 0.97 for the calibration and validation sets, respectively (Table 2). FTIRM1 also showed the high precision among the models with close RMSEC and RMSEP values, which were 0.52 and 0.78% m/m, respectively. Moreover, the RSEP value of FTIRM4 was 1.33%. The worst model in relation to accuracy and precision among the FTIR models was FTIRM2. Thus, this model showed very low r2 values which were 0.51 and 0.49 for both the calibration and validation sets, respectively. Moreover, the model showed high RMSEC, RMSEP and RSEP values of 11.2% m/m, 11.6% m/m and 29.8%, respectively. This indicated that although the model was repeatable, it had low precision as the error values were not satisfactory. Similarly, FTIRM3 (modafinil excipients dilution) showed close RMSEC and RMSEP values of 6.57 and 4.63% m/m, respectively; yet, high RSEP value of 13.55%. FTIRM3 showed low accuracy of calibration with r2 value of 0.75. The same pattern was observed with FTIRM4 that had close values of RMSEC (6.29% m/m) and RMSEP (7.03% m/m) and high RSEP value (19.8%). The lower accuracy in models based on mixtures rather than tablets could be attributed to the small amount of measurements (few milligrams) taken per sample. In this respect, the higher the complexity of the sample (as the case of powdered tablets), the more representation of the sample was in the FTIR spectrum.
NIR models showed the highest accuracy for NIRM3 (modafinil excipients sequential addition) which gave r2 values for the calibration and validation sets of 0.99 and 0.99, respectively (Table 3). NIRM3 showed high precision with RMSEC and RMSEP values of 1.15 and 1.21% m/m correspondingly. Moreover, it showed an RSEP value of 3.45%. Additionally, the two tablet-based models showed high precision but slightly lower accuracy than NIRM3. These included NIRM4 (modafinil powdered tablet model) and NIRM5 (modafinil intact tablet model) which had r2 values of calibration of 0.77 and 0.69 individually. Both models were highly precise and showed RMSEC values below 2% m/m and RSEP values below 4%. The remaining two powder models (NIRM1 and NIRM2) showed slightly lower accuracy but very poor precision. Thus, the r2 values of calibration for NIRM1 (modafinil lactose dilution) and NIRM2 (modafinil excipients dilution) were 0.72 and 0.84. Both of these models showed good repeatability with very close RMSEC and RMSEP values. Thus, NIRM1 showed RMSEC and RMSEP values of 8.45 and 8.82% m/m, respectively. Likewise, NIRM2 showed RMSEC and RMSEP values of 5.26 and 5.25% m/m but had very poor external precision in the range of 15–23%.
Raman models showed the highest accuracy/precision for RamanM4 (modafinil powdered tablet model) and Raman M5 (modafinil intact tablet model) (Table 3). The aforementioned two models showed r2 value of calibration of 0.98 and 0.94. In addition, the RMSEC and RMSEP values for RamanM4 were 0.54 and 0.82% m/m, whereas for RamanM5, these values were 0.96 and 0.91% m/m, respectively. RamanM4 provided a more precise model as it showed ten times lower RSEP value (1.4%) than RamanM5 (12.1%). The models based on powdered mixtures gave lower accuracy and precision than tablet-based models. In this sense, RamanM1 (modafinil lactose dilution), RamanM2 (modafinil excipients dilution) and RamanM3 (modafinil excipients sequential addition) had low r2 values of calibration which were 0.70, 0.84 and 0.76, respectively. The three aforementioned models had high RSEP values which were in the range of 17–24%. All three models showed close agreement between their RMSEC and RMSEP values (Table 3).
Prediction of modafinil in generic tablets
Test sets of powdered and intact tablets were used to examine the external predictive ability of the models. The predicted value was converted into label claim, and the percentage label claim of each product was assessed. The pharmacopoeia acceptable deviation of the API for tablets is usually ± 5% of the label claim in order to allow variation in production, degradation during shelf life of the product and accuracy of the analytical method. In this work, the range was extended to ± 30% of the label claim to compensate for difficulty in setting up a calibration in the spectra and account for the noise generated by the instrument/spectral algorithms (Young-Powell and Page 2014).
For powdered tablets, all the eight products were predicted through the powdered dilution models (Table 4). In this respect, the best predictive ability was observed for NIRM3 which showed a mean prediction of 98.2% label claim (RSD = 2.35%) for all batches. This was followed by FTIRM3 and NIRM1, which showed mean prediction values of 97.9 and 97.2% label claim, respectively. Additionally, FTIRM2 and NIRM2 showed good predictive ability with values of 91.1 and 90.2% label claim, respectively. The remaining models (FTIRM1, RamanM1, RamanM2 and RamanM3) exhibited poor predictive ability below 70%.
RamanM2 and RamanM3 showed better prediction for intact tablets (Table 5). Thus, the mean prediction of intact tablets using the two models were 108 and 84% label claim, respectively. Moreover, NIRM2 showed good predictive ability for intact tablets with a mean prediction of 103% label claim. The remaining models included NIRM1, NIRM3 and RamanM1 had poor prediction above 130% label claim.
The findings demonstrated that the combination of handheld FTIR, NIR and Raman spectroscopy with PLSR offered a rapid method for quantifying modafinil in branded generic medicines with minimal sample preparation. NIR and Raman techniques were non-destructive; however, FTIR required powdering the tablets prior to measurement. In comparison to NIR, FTIR and Raman showed that models based on tablets were more accurate than those based on powder mixtures. Among the powder mixture models, modafinil excipients sequential addition offered the highest accuracy and precision for the quantification of powdered tablets using FTIR and NIR spectroscopy. Modafinil excipient dilution models offered the highest accuracy and precision for the quantification of intact tablets using NIR and Raman spectroscopy. Consequently, the choice of the powder model depended to a degree, on the technique used as well as the sample quantified. Subsequently, this may represent a challenge in the generalisability of the method to other nootropics that could be of different concentration and have different formulation. Hence, future work should consider the accuracy of quantification for different formulation types (tablets, capsules, caplets) and/or closely related analogues of drugs.
Availability of data and materials
The datasets used and/or analysed during the current study are available from the corresponding author on reasonable request.
Active pharmaceutical ingredient
Correlation in wavelength space
High-performance liquid chromatography
Partial least square regression
Relative standard error of calibration
Relative standard error of prediction
Root mean square error of calibration
Thin layer chromatography
World Health Organisation
Assi S, Watt RA, Moffat AC. Identification of counterfeit medicines from the Internet and the World market using near-infrared spectroscopy. Analytical Methods. 2011a;3(10):2231–6.
Assi S, Watt RA, Moffat AC. Authentication of medicines using Raman spectroscopy. Eur Pharm Rev. 2011b;16(1):49–55.
Assi S, Watt RA, Moffat AC. On the quantification of ciprofloxacin in proprietary Ciproxin tablets and generic ciprofloxacin tablets using handheld Raman spectroscopy. Journal of Raman Spectroscopy. 2012;43(8):1049–57.
Baskys, A.; Remington, G., Brain mechanisms and psychotropic drugs. CRC Press 1996, 35.
Bouyé, C.; Robin, T.; d'Humières, B., Spectral imaging spreads into new industrial and on-field applications. SPIE 2018, 10539.
Brereton, R.G., Chemometrics: data analysis for the laboratory and chemical plant. John Wiley & Sons 2003.
Burns, D.A. and Ciurczak, E.W. Handbook of near-infrared analysis. CRC press 2007.
Caporaso N, Whitworth MB, Fisk ID. Near-Infrared spectroscopy and hyperspectral imaging for non-destructive quality assessment of cereal grains. Appl Spectrosc Rev. 2018;53(8):667–87.
Cass Q, Kohn C, Calafatti S, Aboul-Enein H. An enantioselective assay for (±)-modafinil. J Pharm Biomed Anal. 2001;26(1):123–30.
Champagne J, Gardner B, Dommett EJ. Modelling predictors of UK undergraduates’ attitudes towards smart drugs. Trends in neuroscience and education. 2019;2019(14):33–9.
Correia RM, Domingos E, Tosato F, dos Santos NA, Leite JDA, da Silva M, Marcelo MCA, Ortiz RS, Filgueiras PR, Romão W. Portable near infrared spectroscopy applied to abuse drugs and medicine analyses. Anal Methods. 2018;10(6):593–603.
Crocombe RA. Portable spectroscopy. Applied Spectroscopy. 2018;72(12):1701–51.
Faisall M, Naz Z, Shakeel F, Ahmed S, Kohli K, Khar R. A New TLC densitometric method for stability assessment of modafinil. Chemia Analityczna. 2009;54(1):77–88.
FDA The possible dangers of buying medicines over the Internet; 07-August-2014.
Fedchak, S. Presumptive field testing using portable raman spectroscopy; Las Vegas Metropolitan Police Department, USA 2014, 1-59.
Fittler A, Bősze G, Botz L. Evaluating aspects of online medication safety in long-term follow-up of 136 Internet pharmacies: illegal rogue online pharmacies flourish and are long-lived. Journal of medical Internet research. 2013;15(9):E199.
Gaurvika MLN, Breman JG, Mackey TK, Clark JP, Hajjou M, Littrell M, Herrington JE. Falsified and substandard drugs: stopping the pandemic. Am J Trop Med Hyg. 2019;100(5):1058–65.
Gerace E, Seganti F, Luciano C, Lombardo T, Di Corcia D, Teifel H, Vincenti M, Salomone A. On-site identification of psychoactive drugs by portable Raman spectroscopy during drug-checking service in electronic music events. Drug and alcohol review. 2019;38(1):50–6.
Gold LH, Balster RL. Evaluation of the cocaine-like discriminative stimulus effects and reinforcing effects of modafinil. Psychopharmacology. 1996;126(4):286–92.
International Narcotics Control Board. Availability of internationally controlled drugs: ensuring adequate access for medical and scientific purposes: indispensable, adequately available and not unduly restricted. UN 2016.
Janvier S, Spiegeleer BD, Vanhee C, Deconinck E. Falsification of biotechnology drugs: current dangers and/or future disasters? J Pharm Biomed Anal. 2018;161:175–91.
Jee RD. Near-infrared Spectroscopy. In: Moffat, Anthony C; Osselton, M David; Widdop, Brian; Watts, Jo. Clarke’s analysis of drugs and poisons 2019.
Moffat AC, Assi S, Watt RA. Identifying counterfeit medicines using near infrared spectroscopy. J Near Infrared Spectrosc. 2009;18(1):1–15.
Mukhopadhyay R. The hunt for counterfeit medicine. Analytical Chemistry. 2007;79(7):2622–7.
Nageswara Rao R, Dhananjay D, Kumar TM. Development and validation of a reversed phase liquid chromatographic method for separation and determination of related substances of modafinil in bulk drugs. Talanta. 2007;73(3):407–14.
Orizio G, Merla A, Schulz PJ, Gelatti U. Quality of online pharmacies and websites selling prescription drugs: a systematic review. Journal of medical Internet research. 2011;13(3):e74.
Orizio G, Schulz P, Domenighini S, Caimi L, Rosati C, Rubinelli S, Gelatti U. Cyberdrugs: A cross-sectional study of online pharmacies characteristics. European Journal of Public Health. 2009;19(4):375–7.
Rao RN, Shinde DD, Talluri MK. Development and validation of a reversed phase liquid chromatographic method for separation and determination of related-substances of modafinil in bulk drugs. Talanta. 2007;73(3):407–14.
Safemedicines INTERPOL-Led sting operation shuts down over 9,600 illegal Internet pharmacies, seizes $41 million in fake, dangerous or unapproved drugs; July-2013.
The Medicines Compendium Modafinil Provigil 100 mg Tablets; February 2020, 2020. https://www.medicines.org.uk/emc/medicine/28926.
Turner DC, Robbins TW, Clark L, Aron AR, Dowson J, Sahakian BJ. Cognitive enhancing effects of modafinil in healthy volunteers. Psychopharmacology. 2003;165(3):260–9.
Venkatesh V, Lakshmi NV, Suresh PV, Rao PM, Siva K, Raju GD, Rao NR. Determination and validation of Modafinil in Pharmaceutical Formulation by Spectrophotometric and RP- HPLC Methods. J Pharm Res. 2011;4(2):509–11.
WHO Counterfeit medicines; Switzerland, 14-November-2006, 2006.
WHO Medicines: spurious/falsely-labelled/ falsified/counterfeit (SFFC) medicines; WHO: Switzerland, May-2012, 2012.
Xu XP, Wang S, He YJ. HPLC Determination of modafinil and its capsules. Chinese J Pharma Anal. 2002;22:191–2.
Yang J, Lou X, Yang H, Yang H, Liu C, Wu J, Bin J. Improved calibration transfer between near-Infrared (NIR) spectrometers using canonical correlation analysis. Analytical Letters. 2019;52(14):2188–202.
Young-Powell A, Page L. One in five students have taken the study drug modafinil. The Guardian; May-2014, 2014.
The authors would like to than Bruker UK Ltd. for the mobile IR instrument, and SciMed for the FirstGuard Raman and microNIR instruments.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Assi, S., Khan, I., Edwards, A. et al. On-spot quantification of modafinil in generic medicines purchased from the Internet using handheld Fourier transform-infrared, near-infrared and Raman spectroscopy. J Anal Sci Technol 11, 35 (2020). https://doi.org/10.1186/s40543-020-00229-3
- Counterfeit medicines
- Partial least square regression