- Short report
- Open Access
Detection of cathinone and mephedrone in plasma by LC-MS/MS using standard addition quantification technique
Journal of Analytical Science and Technology volume 8, Article number: 19 (2017)
Cathinones better known as bath salts have been listed as illicit drugs since 2011. Few studies have focused on the analytical extraction techniques and the matrix effect affecting their detection and quantification in biological samples. Matrix suppression of the signal of cathinones has been previously observed in urine sample by LC-MS/MS. This study is aimed to use the standard addition method to overcome the plasma matrix effect on the quantification of cathinone and mephedrone by LC-MS/MS.
The results showed the matrix effect for cathinone at lowest tested concentration (10 ng/ml) was significantly reduced from 210.9 to 133.7%, but not for mephedrone (from 196.8 to 191.9%) by using standard addition method. At the higher tested concentrations of samples, the matrix effects were significantly reduced for both cathinone and mephedrone by using standard addition technique.
Standard addition quantitative technique can serve as an alternative quantitative method for LC-MS/MS when suitable internal standards are not available.
Cathinones are also known as bath salts and have been listed as illicit drugs since 2011. The Administrator of the Drug Enforcement Administration (DEA) issued the order to temporarily schedule three synthetic cathinones as schedule I substances under the Controlled Substances Act (CSA) (Drug Enforcement Administrations 2011; 2014). Internationally, cathinones have been banned in numerous other countries. Designer cathinones, which are newly synthesized derivatives of existing cathinones to mimic the effects of existing illegal drugs and at the same time bypass the existing controlled drug regulations, are still available (Kelly 2011). These products have caught the attention of law enforcement due to incidents of acute toxicity and numerous fatalities that have been linked with the use of these designer cathinones.
Accurately, quantifying the concentration of cathinones in biological samples is important for the regulation of cathinones. A problem for law enforcement and clinical laboratories occurs due to the difficulties of validation and maintenance of multiple analytical methods used in the accurate and precise detection of these compounds. Despite the increased availability of designer drugs, few studies have focused on the analytical extraction techniques and the matrix effects affecting their detection and quantification in biological samples.
LC-MS/MS is commonly used for detecting cathinones. However, Concheiro et al. (2013) showed the matrix suppression was observed in urine sample of cathinones by LC-MS/MS analysis. Endogenous interferences in urine have suppressed cathinone, mephedrone, and 3,4-methylenedioxypyrovalerone (MDPV) detections by LC-MS/MS up to 27, 11, and 9%, respectively. By IUPAC definition, matrix effects are “the combined effect of all components of the sample other than the analyte on the measurement of the quantity” (Guilbault and Hjelm 1989). Many studies have been conducted to provide the solution to overcome the matrix effects (Lehotay et al. 2010; Stahnke et al. 2012). This study was aimed to explore the standard addition method as a means to overcome the plasma matrix effects on the quantification of cathinone and mephedrone by LC-MS/MS. We developed LC-MS/MS methods that utilized standard addition without internal standard (IS).
There are three commonly used techniques used to quantify the unknown samples: (1) external calibration with solvent standards, (2) external calibration with matrix matched standards, and (3) standard addition. Internal standard (IS) is important for using the external calibration to determine the chemical concentration. Isotopically labeled IS is the ideal standard to overcome the matrix effects and accurately quantify the known chemicals in the samples. When isotopically labeled IS is not available, a non-isotopically labeled IS can be used. However, the non-isotopically labeled IS should not be present in any of the tested samples and also should be as similar in physiochemical properties of the analyte. But these ideal internal standards are not always available especially for those new designer drugs. Standard addition (post-extraction spike) has been shown to compensate for the matrix effects contributed by biological matrices (Van Eeckhaut et al. 2009). Standard addition is commonly used for atomic absorption spectroscopy and gas chromatography analyses (Bonilla 1978; Wurita et al. 2014). However, there are few studies using the standard addition method for mass spectroscopy (Hasegawa and Suzuki 2014; Zenkevich and Morozova 2014). This study used the standard addition technique to determine cathinone and mephedrone concentrations in plasma by LC-MS/MS.
he results showed the matrix effect for cathinone at lowest tested concentration (10 ng/ml) was significantly reduced from 210.9 to 133.7%, but not for mephedrone (from 196.8 to 191.9%) by using standard addition method. At the higher tested concentrations of samples, the matrix effects were significantly reduced for both cathinone and mephedrone by using standard addition technique. Standard addition quantitation technique can be used to overcome the matrix effect when appropriate internal standard is not available.
Materials and methods
Solvents, cathinone (1 mg/ml in methanol), and mephedrone (1 mg/ml in methanol) were purchased from Sigma-Aldrich (St. Louis, MO, USA). Sheep plasma was obtained from Carolina (Burlington, NC). Cathinone and mephedrone were diluted either in methanol as “solvent standard” or in plasma with known concentrations (0, 10, 33, and 100 ng/ml) of drugs as “standard addition” to at least five different calibration points (0–250 ng/ml). Solid phase extraction (SPE) was performed using Strat-X™ Drug B cartridges (Phenomenex).
Solid phase extraction
Plasma samples (1 ml) were mixed with 1 ml of 1 M acetic acid, then centrifuged. After centrifugation, the supernatant was subjected for SPE extraction. SPE cartridges were conditioned with 2 ml of methanol and then 2 ml of water. After loading sample, the SPE cartridges were washed separately with 2 ml of 1 M acetic acid and then 2 ml of methanol. Two milliliters of elution solvent (methylene chloride to isopropanol to ammonium hydroxide = 80:18:2) was passed through the SPE column and collected. After extraction, SPE extracted plasma with cathinone and mephedrone samples were either analyzed directly by LC-MS/MS or first spiked with known concentrations (0–250 ng/ml) of cathinone and mephedrone as “standard addition” samples. Two hundred (200) microliters of HCl in methanol (2%) was added to samples before drying with a Turbovap N2 evaporator. After drying, 200 ul of mobile phase A was used to reconstituent samples for LC-MS/MS analysis.
UPLC-MS/MS (Shimadzu LCMS 8030) with a Phenomenex Kinetex™ C18 column (2.1 × 100 mm, 1.7 um) was used to separate compounds under the following condition: gradient mobile phase system (A: 0.1% formic acid in water to methanol and B: 0.1% formic acid in acetonitrile; 0 min 5% B, 3 min 40% B, 3.5 min 90% B, 4 min 5% B); column heater: 30 °C; 10 ul injection volume; flow rate 0.5 ml/min). The Shimadzu LCMS 8030 triple quad instrument was used with the following settings: 1.5 L/min nebulizing gas flow, 250 °C DL temperature; 400 °C heat block temperature; 15 L/min drying gas flow). Mass transitions are as following: cathinone 150.1 > 117.0 (Q1–13, CE – 23, Q3–30); 150.1 > 104.85 (Q1–13, CE – 23, Q3–27) and mephedrone 178.0 > 145.0 (Q1–13, CE – 23, Q3–14); 178.0 > 90.95 (Q1–15, CE – 37, Q3–17).
Results and discussion
Plasma samples containing cathinone and mephedrone (10, 100, and 500 ng/ml) were extracted using Strat-X™ Drug B cartridge and analyzed by LC-MS/MS together with solvent standards. The matrix effects on the extracted samples were significantly observed in low concentration (10 ng/ml) samples (Fig. 1). Since the cutoff concentration for cathinone and mephedrone is 25 ng/ml, the matrix effect will make a possible false positive determination. In order to overcome this matrix effect, extracted plasma samples with cathinone and mephedrone (0, 10, 33, and 100 ng/ml) were spiked with various concentrations of standards (0–250 ng/ml) and analyzed by LC-MS/MS. Calibration curves were established and shown in Figs. 2 and 3 (A solvent standard, B standard addition) for cathinone and mephedrone, respectively.
Concentrations of cathinone and mephedrone were then calculated by subtracting the value of X-axis intercept of solvent standard calibration from the value of X-axis intercept of standard addition calibration (Tables 1 and 2). The results show the matrix effect for cathinone at 10 ng/ml level was significantly reduced from 210.9 to 137.7%, but not for mephedrone (196.8 vs 191.9%). At higher concentration samples, the matrix effect for both cathinone and mephedrone was not significant compared with 10 ng/ml samples (Table 3).
The study here using standard addition quantitative technique showed a significant reduction of plasma matrix effect on cathinones analysis by LC-MS/MS. However, this standard addition technique for quantifying plasma samples by LC-MS/MS requires sufficient amount of sample to set the standard addition calibration curve and more work (two calibration curves) is needed. Therefore, this technique can serve as an alternative quantitative method to overcome the matrix effect when no appropriate internal standard is available.
Drug Enforcement Administration
Liquid chromatography tandem mass spectrometry
Solid phase extraction
Bonilla E. Flameless atomic absorption spectrophotometric determination of manganese in rat brain and other tissues. Clin Chem. 1978;24:471–4.
Concheiro M, Anizan S, Ellefsen K, Huestis MA. Simultaneous quantificaiton of 28 synthetic cathinones and metabolites in urine by liquid chromatography-high resolution mass spectrometry. Anal Bioanal Chem. 2013;405:9437–48.
Drug Enforcement Administrations. 21 CFR part 1308. Schedules of controlled substances: temporary placement of three synthetic cathinones into schedule I. Fed Regist. 2011;76:65371–5.
Drug Enforcement Administrations. 21 CFR part 1308. Schedules of controlled substances: temporary placement of 10 synthetic cathinones into schedule I. Fed Regist. 2014;79:12938–43.
Guilbault GG, Hjelm M. Commissions on analytical nomenclature and automation and clinical chemical techniques. Pure Appl Chem. 1989;61:1657–64.
Hasegawa K, Suzuki O. Postmortem distribution of α-pyrrolidinovalerophenone and its metabolite in body fluids and solid tissues in a fatal poisoning case measured by LC-MS-MS with the standard addition method. Forensic Toxicol 2014;32:225-234.
Kelly JP. Cathinone derivatives: a review of their chemistry, pharmacology and toxicology. Drug Test Anal. 2011;3:439–53.
Lehotay SJ, Mastovska K, Lightfield AR, Gates RA. Multi-analyst, multi-matrix performance of the QuEChERS approach for pesticide residues in foods and feeds using HPLC/MS/MS analysis with different calibration techniques. J AOAC Int. 2010;93:355–67.
Stahnke H, Kittlaus S, Gunther K, Alder L. Reduction of matrix effects in liquid chromatography-electrospray ionization-mass spectrometry by dilution of the sample extracts: how much dilution is needed? Anal Chem. 2012;84:1474–82.
Van Eeckhaut A, Lanckmans K, Sarre S, Smolders L, Michotte Y. Validation of bioanalytical LC-MS/MS assays: evaluation of matrix effects. J Chromatogr B. 2009;877:2198–207.
Wurita A, Hasegawa K, Minakata K, Gonmori K, Nozawa H, Yamagishi I, Suzuki O, Watanabe K. Postmortem distribution of α-pyrrolidinobutiophenone in body fluids and solid tissues of a human cadaver. Legal Med 2014;16:241-246.
Zenkevich IG, Morozova TE. Areas of application and characteristics of quantitative chromatographic analysis by the consecutive standard addition method. J Anal Chem. 2014;69:327–35.
We specially thank Argeliz Pomales and Teeshavi Narayne for excellent technical assistance and Scott A. Cassidy and William Orlando from Shimadzu for guidance and support.
The work is supported by the PRISM fund at CUNY John Jay College. The PRISM is the Program for Research Initiatives for Science Majors at John Jay College and funded by the Title V, HSI-STEM, and MSEIP programs within the US Department of Education; the PAESMEM program through the National Science Foundation; and New York State’s Graduate Research and Teaching Initiative.
Availability of data and materials
The data of this project can be shared.
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Cheng, SY., Ng-A-Qui, T., Eng, B. et al. Detection of cathinone and mephedrone in plasma by LC-MS/MS using standard addition quantification technique. J Anal Sci Technol 8, 19 (2017). https://doi.org/10.1186/s40543-017-0128-7
- Standard addition