- Technical note
- Open Access
Stability study of opioids and benzodiazepines in urine samples by liquid chromatography tandem mass spectrometry
© Dixon et al.; licensee Springer. 2015
- Received: 11 November 2014
- Accepted: 27 April 2015
- Published: 17 May 2015
The stability of opioids and benzodiazepines was studied in patient urine samples stored at different temperatures over 30 days.
Samples were prepared by solid phase extraction and quantified by liquid chromatography coupled to a tandem triple quadrupole mass spectrometer.
Sample storage conditions affected the stability of all drugs investigated. The concentrations of hydromorphone and oxymorphone in urine samples from patients who had taken hydrocodone, morphine, and/or oxycodone increased significantly over 30 days of storage.
These data suggest that results from long-term-stored urine samples with opioids and benzodiazepines should be cautiously interpreted. This study indicates that the biotransformation of parent drugs to metabolites still occurs during sample storage at ambient temperatures.
- Solid phase extraction
- Triple quadrupole mass spectrometry
- Liquid chromatography
In the USA, more attention has been placed during the past decade on effective pain management due to its contribution to the healing process and the improvement of the quality of patients’ lives. However, according to the US Centers for Disease Control and Prevention, pain medication prescription overdoses have increased rapidly and are now at epidemic levels (Centers for Disease Control and Prevention 2012). Recent studies have shown that physicians, internists, and family practitioners are more likely than ever before to be supervising care of patients with chronic pain who regularly have other comorbidities and may be taking medications that could affect the results of urine drug test (Couto et al. 2009). Couto et al. found in their recent article that three quarters of patients from nearly one million patient test samples were not taking their medications according to their physician’s prescription. Altogether, 38% of patients did not have a detectable level of their prescribed medication, while 27% had higher drug levels and 15% had lower drug levels than expected (Couto et al. 2009).
A review on the topic of drug instability by Peters indicates that storage conditions are important factors for data interpretation. Of particular note, compounds that are structurally labile are most susceptible to degradation (Peters 2007). Benzodiazepines and opioids are among the most commonly prescribed drugs in pain management therapy. Benzodiazepines are prescribed particularly in anxiety-induced depression cases, stress, panic, sleep disorders, and seizures, while opioids have long been used to treat acute pain such as postoperative pain and to alleviate the severe and disabling pain of terminal conditions such as cancer. However, these substances are regularly present in emergency intoxication episodes and drugs-of-abuse testing. They are also regularly responsible for overdose in suicide attempts (Lader 1999; Drummer 2004). Consequently, clinical and forensic laboratories must be able to accurately screen, confirm, and quantify benzodiazepines and opioids along with their metabolites in biological samples.
The knowledge on drug concentration changes in the sample and on the time of its storage period is crucial particularly if there is a considerable time interval between sample collection and toxicological analysis (Skopp 2004). Samples are usually kept at undefined storage conditions during transportation to the laboratory, and toxicology laboratories are obliged to keep biological samples for a certain period in order to allow reanalysis if it is required by the practitioner. Hence, the instability of the drugs (or benzodiazepines and opiates in this case) in different biological samples (or urine in this case) is a problem of data interpretation in analytical toxicology. Biological samples for toxicological analysis frequently remain at room temperature for some hours during transport, reception, and registration. Refrigeration of samples at 4°C is common following drug screening and analysis confirmation. For forensic purposes, samples may be frozen for long time periods due to legal constraints and subsequent retest. Some factors such as storage time and temperature can influence drug stability. It is one of the most important issues challenging the interpretation of postmortem toxicology results (Drummer 2007). A number of studies in the literature are focused on the effects of sample storage temperature and the formation of metabolites (El Mahjoub and Staub 2000; Skopp et al. 2001a; Skopp et al. 2001b; Klingmann et al. 2001; Clauwaert et al. 2001), but these mainly report data from blood and/or plasma samples.
Due to the high number of intoxications per year with drugs and pharmaceuticals, fast drug screening methods are necessary in clinical and forensic intoxication cases. Traditional methods usually include immunoassay testing and gas chromatography/mass spectrometry (GC/MS) for confirmation. There are intrinsic strengths and weaknesses associated with these methods. Immunoassays are only available for a small number of substance classes, and it suffers from lack of specificity and sensitivity (Hammett-Stabler and Cotten 2012). GC/MS requires a rigorous sample preparation often involving derivatization of certain compounds and relatively long chromatographic runs (Anizan et al. 2012). Liquid chromatography tandem mass spectrometry (LC-MS/MS) has been increasingly used for confirmation of drugs of abuse in various matrices (de Castro et al. 2009a; de Castro et al. 2009b; Cheng et al. 2006; Mueller et al. 2005; de Jager and Bailey 2011). Multiplex triple quadrupole mass spectrometry has enabled simultaneous detection of a drug and its fragment ions in one single run.
The aim of the present study was to evaluate the stability of opioids and benzodiazepines in patient urine samples after different storage periods and temperatures. A triple quadrupole mass spectrometer was used to monitor the stability of six opioids and five benzodiazepines in urine samples from four patients with chronic pain, who have been under physicians’ supervision. Patient I was prescribed morphine and oxycodone (Oxy Ir®), while patient II was prescribed oxymorphone (Opana®) and oxycodone. Patient III was prescribed hydrocodone and acetaminophen (Lortab®), oxycodone and acetaminophen (Percocet®), carisoprodol (Soma®), and diazepam (Valium®), and patient IV was prescribed oxycodone and acetaminophen (Endocet®), carisoprodol (Soma®), and alprazolam (Xanax®). The stability of parent compounds and corresponding metabolites was evaluated in urine samples stored at laboratory temperature or in the refrigerator over 30 days.
Materials and reagents
Certified reference standards 6-acetylmorphine, morphine, buprenorphine, hydromorphone, norbuprenorphine, hydrocodone, oxycodone, oxymorphone, alpha hydroxy alprazolam, alprazolam, nordiazepam, oxazepam, and temazepam were purchased from Cerilliant (Round Rock, TX, USA). For quality control samples, hydromorphone, oxycodone, oxymorphone, alprazolam, oxazepam, and temazepam were purchased from Restek (Bellefonte, PA, USA) while buprenorphine, alpha hydroxy alprazolam, and nordiazepam were purchased from Grace (Deerfield, IL, USA). Norbuprenorphine, 6-acetylmorphine, and alpha hydroxy alprazolam quality control samples were different lot numbers from Cerilliant. Deuterated internal standards oxycodone-d3, buprenorphine-d4, hydromorphone-d3, norbuprenorphine-d3, oxymorphone-d3, alpha hydroxy alprazolam-d5, alprazolam-d5, oxazepam-d5, and temazepam-d5 were purchased from Cerilliant. Each standard compound (1 mg/mL) and internal standard (0.1 mg/mL) in methanol was prepared in 30% methanol as working solutions. Acetonitrile, methanol, and water were LC-MS grade and purchased from J.T. Baker (Phillipsburg, NJ, USA). Reagent-grade formic acid, phosphate buffer, and β-glucuronidase enzyme were purchased from Sigma-Aldrich (St. Louis, MO, USA). Ammonium formate was purchased from Hampton Research (Aliso Viejo, CA, USA). The 96-deep-well plates for solid phase extraction (SPE; GV-65C 96-deep-well) and sample collection were purchased from Scienceware (Wayne, NJ, USA), while the Agilent 96-well plate (Agilent Technologies, Santa Clara, CA, USA) was used on the LC-MS autosampler.
Sample collection and storage
Urine samples were obtained from four patients (I to IV) with prescribed medications containing opioids and/or benzodiazepines. Samples were received in our laboratory within 24 h after collection and the analyses were performed immediately. For compound stability investigation, urine samples were separated into two aliquots, in polypropylene tubes, and stored at room temperature (about 20°C) and at 4°C for 30 days. Urine samples were prepared by SPE and analyzed by liquid chromatography tandem mass spectrometry.
Sample preparation and SPE extraction
Samples were prepared using a Tecan Genesis RSP 100 Liquid Handler (Tecan Group Ltd., Männedorf, Switzerland). For the standard calibration curve, 100 μL of a 100-μg/mL stock solution (final concentration 5 μg/mL) of internal standards was combined to 30% methanol in water including the reference standard. The final volume of each calibration standard level (1 to 6) was 2 mL. For patient urine samples, 150 μL of urine samples, 38 μL of internal standard mixed (250 ng/mL final concentrations), 300 μL of 0.1 M phosphate buffer pH 6.8, 130 μL of β-glucuronidase enzyme, and 108 μL of water were added in a 96-deep-well plate. An incubation at 47°C for 2 h on a shaker followed. Following the enzymatic digestion, 24 μL of 10% ammonium hydroxide was added to each well and the plate was left at room temperature for 20 min.
For SPE, the 96-deep-well SPE plate on a vacuum manifold was preconditioned with methanol and 250 mM phosphate buffer pH 9. Samples (500 μL) were then added, washed twice with 200 μL of 250 mM phosphate buffer pH 9, and eluted with three additions of 400 μL methanol. The eluate (1,200 μL) was evaporated under nitrogen using a 96-well-plate evaporator system. Samples were then reconstituted in 500 μL of 30% methanol in water, incubated on a shaker for 10 min, and centrifuged at 3,000 rpm for 10 min. The supernatant (250 μL) was then transferred to the Agilent 96-well plate for LC-MS analysis.
LC-MS analyses were performed on an Agilent 1290 Infinity LC coupled to an Agilent 6410 triple quadrupole mass spectrometer (Agilent Technologies, Santa Clara, CA, USA) with an electrospray ionization source. Chromatographic separation was carried out in a Poroshell 120SB-C18 column (2.1 × 100 mm, 2.7 μm; Agilent Technologies, Santa Clara, CA, USA) with 99.9% water, 0.05% formic acid, and 0.05% ammonium formate as mobile phase A, and 99.95% acetonitrile and 0.05% formic acid as mobile phase B. The autosampler and the column temperature were maintained at 6°C and 50°C respectively. Gradient elution with mobile phase A and mobile phase B at a flow rate of 0.7 mL/min was as follows: 5% to 50% B over 3 min, 50% to 95% B over 0.5 min, isocratic at 95% B for 0.6 min, 5% to 95% B over 0.1 min, and isocratic at 5% B over 0.7 min. Prior to the next injection, the column was re-equilibrated at 5% B over 1 min. A divert valve switched the LC flow initially to waste for 0.3 min, subsequently to the mass spectrometer for 4.5 min, and then back to the waste for the remaining chromatographic run.
Multiple reaction monitoring (MRM) transitions with optimized fragmentation voltage and collision energy
MRM transition ( m / z )
328.2 > 165.1
328.2 > 211.1
468.3 > 55.1
468.3 > 396.2
286.2 > 185.1
286.2 > 128.1
414.3 > 152.1
414.3 > 165.1
316.2 > 241.1
316.2 > 256.1
302.1 > 284.2
302.1 > 227.1
286.2 > 165.1
286.2 > 157.1
300.2 > 199.1
300.2 > 171.1
Alpha hydroxy alprazolam
Alpha hydroxy alprazolam-d5
325.1 > 216.1
325.1 > 279.1
309.1 > 281.1
309.1 > 205.1
271.1 > 140.1
271.1 > 208.1
287.1 > 241.1
287.1 > 269.1
301.1 > 255.1
301.1 > 177.1
Selectivity, sensitivity, linearity, intra-assay imprecision and analytical recovery, extraction efficiency, matrix effect, process efficiency, hydrolysis, and drug carry over were evaluated. Potential endogenous interferences were determined by the analysis of blank urine samples from laboratory volunteers. Sensitivity of the method was examined by establishing the limit of quantitation (LOQ), the half limit of quantitation (1/2 LOQ), and the introduction of the initial calibration verification standards (ICV) in the calibration curve standards. The accuracy of each compound at each calibration level was between 80% and 120%. Linearity was determined using quadratic regression with two replicates for each level. Six-level calibration curves were used with the dilution pattern 1:2:2:2:2:4 for all compounds.
Extraction efficiency, matrix effect, and process efficiency were calculated at three QC concentrations. A set of samples containing the same amount of internal standard and blank urine were spiked with different amounts of analytes corresponding to levels 1, 3, and 5 of the calibration curve. The matrix effect was assessed by comparing the peak areas of the analyte standards in matrix-free solvent with standards spiked into blank urine samples after extraction. The recovery and ion suppression or enhancement associated with the blank urine was evaluated by dividing the concentration of standards from matrix-free with standards in blank urine multiplied by 100.
The combination of SPE followed by LC-MS analysis is routinely used for metabolomics (Cheng and Mok 2004; Lu et al. 2008; Zhou et al. 2012). A similar procedure has been used to successfully quantify opioids and benzodiazepines from patient urine samples, using a triple quadrupole mass spectrometer. Triple quadrupole tandem mass spectrometry provides sensitive quantitation by acquiring the signal corresponding to the highest response product ion (quantifier) from the fragmentation of the compound parent ion or precursor ion. The transition between the parent ion and the product ion, known as MRM, is subsequently used to acquire an additional signal corresponding to the next highest product ion (qualifier) (Table 1) (Yost and Enke 1979; Yao et al. 2010). This information may be considered as adequate for confirmation and quantitation of analyte, particularly if the ratio of signal between the two product ions is consistent between the calibration standards and the unknown samples. Dynamic MRM enables MRM acquisitions of compounds during a retention time window corresponding to the elution time of each compound, thus increasing the minimum dwell time (Liu et al. 2011).
Concentration of opioids and benzodiazepines in patient urine samples
Detected compounds and metabolites
Alpha hydroxyl alprazolam
Lortab®, Percocet®, Soma®, Valium®
Hydrocodone, acetaminophen, oxycodone, carisoprodol, diazepam
Alpha hydroxyl alprazolam
Endocet®, Soma®, Xanax®
Oxycodone, acetaminophen, carisoprodo, alprazolam
Alpha hydroxyl alprazolam
Stability of opioids
6-Acetylmorphine, a biologically specific indicator marker of heroin, has a high occurrence in oral fluid from users of this drug, but due to its short half-life, its detection in urine is limited (Allen 2011). Less than 50% of this metabolite remained in the patient urine sample after 3 days (Figure 3), and its concentration remained nearly constant for the rest of the storage duration. After ingestion of heroin, it breaks down in minutes to 6-acetylmorphine and then to morphine. This drug is not medically prescribed in the USA, and it has been given schedule 1 status by the US Drug Enforcement Administration. This drug has been detected in patient I’s urine sample with prescribed medications containing morphine and oxycodone.
Buprenorphine and its metabolite norbuprenorphine were both detected in patient II. The concentration of the parent compound decreased over 8 days of storage, while the concentration of its metabolite increased (Figure 3). The stability of buprenorphine and norbuprenorphine has recently been investigated in whole blood samples from suspected drug users (Selden et al. 2011). This investigation revealed that buprenorphine and norbuprenorphine showed some degradation when stored at 4°C for 3 weeks, and in many cases, the concentrations of buprenorphine were very low compared to norbuprenorphine. These findings support the evidence that buprenorphine may be converted to norbuprenorphine in whole blood sample at 4°C, but this conversion in urine sample has not previously been demonstrated. In the present study, the increase in the concentration of norbuprenorphine is probably due to degradation of buprenorphine.
Hydromorphone and oxymorphone were detected in patients I, III, and IV who were prescribed medications containing morphine and/or oxycodone. Evidence for morphine metabolism to hydromorphone in pain patients chronically treated with morphine has extensively been confirmed (Cone et al. 2006; Hughes et al. 2012). Likewise, the metabolism of oxycodone to oxymorphone in pain patients has also been demonstrated (Yee et al. 2012). However, the occurrence of this transformation in stored urine samples has not been previously demonstrated. After ingestion, morphine is metabolized in the liver into morphine-3-glucuronide and morphine-6-glucuronide (Kilpatrick and Smith 2005). It may also be metabolized into normorphine, codeine, and hydromorphone, but the metabolism rate is determined by gender, age, diet, genetic makeup, disease state, and the use of other medications. In humans, oxycodone is primarily metabolized in the liver to α and β oxycodol and oxymorphone (Moore et al. 2003). Some patients are fast metabolizers of oxycodone, resulting in reduced analgesic effect but increased adverse effects, while others are slow metabolizers resulting in increased toxicity without improved analgesia (Gasche et al. 2004). The concentrations of the metabolites hydromorphone and oxymorphone in urine samples after 30 days of storage were very high compared to the initial concentration at day 1 and independently to patients. This trend once more supports the evidence that parent compounds may be converted into corresponding metabolites in urine samples stored at 4°C. The metabolite concentration levels depend on the concentration of the parent compounds. These two metabolites can still be analyzed in patient urine samples stored at 4°C for 30 days, when patients have been prescribed medications containing the main compounds.
Stability of benzodiazepines
Unlike the variation in the concentrations of opioids investigated, the concentrations of all benzodiazepines decreased over 30 days of storage at 4°C. Alprazolam, nordiazepam, and oxazepam showed the best stability with 95%, 90% to 98%, and 97% of the original concentration at day 1 respectively, while alpha hydroxy alprazolam (43% to 85%) and temazepam (68% to 78%) showed more degradation (Figure 3).
In humans, alprazolam is extensively metabolized to alpha hydroxy alprazolam and 4-hydroxy alprazolam, whose plasma concentrations relative to unchanged alprazolam concentration are less than 4% (Smith et al. 1984). The stability of alprazolam is attributed to the presence of the triazole ring in its structure which makes the compound more resistant to hydrolysis. In a recent study on the stability of benzodiazepines in postmortem samples, estazolam whose structure is similar to alprazolam was the most stable in unpreserved blood samples at 4°C (Melo et al. 2012). The coefficient of variation ranged from 1.4% to 6% over 4 weeks of storage. Alprazolam and alpha hydroxy alprazolam were detected in patients II and IV, but only patient IV has been prescribed medications containing alprazolam.
The main active metabolite of diazepam is nordiazepam and the minor active metabolites are oxazepam and temazepam (Riss et al. 2008). The half-life of diazepam is about 1 to 3 days and 2 to 7 days for nordiazepam. The elimination half-life of diazepam and nordiazepam increases considerably in the elderly, which may result in the prolonged action and accumulation of the drug during repeated administration (Vozeh 1981). Nordiazepam, oxazepam, and temazepam were detected in patients I and III, but only patient III was under prescription of medications containing diazepam, the parent compound. Nordiazepam and oxazepam were the most stable compared to temazepam. Atanasov et al. recently investigated the stability of diazepam in blood samples at different storage conditions (Atanasov et al. 2012). The authors showed that the concentration of diazepam decreased by 30% after storage for 4 weeks at 4°C. However, diazepam and chlordiazepoxide are benzodiazepines with long-acting active metabolites like nordiazepam. We could also observe an increase in the concentration of nordiazepam (patients I and III) after 2 weeks of storage at 4°C, thus suggesting diazepam degradation into nordiazepam (Figure 3). The data obtained from this study suggested that opioid and benzodiazepine degradation with consequent metabolites may occur by chemical breakdown during storage. In postmortem samples, it has been suggested that decomposition may occur due to microorganisms’ activity (Melo et al. 2012; Levine et al. 1983).
This work implies that some patients do not take their medications according to their physician’s prescription or take illicit drugs in addition to their prescriptions. When interpreting toxicological results related to possible opioid or benzodiazepine exposure, it is important to consider that the presence of their metabolites in urine samples can be due to patient biotransformation of the drug (in vivo), but also due to their decomposition during storage (in vitro). Opioid and benzodiazepine concentrations in urine samples can change due to degradation during sample storage. This study revealed that opioid and benzodiazepine degradation increased with storage time. The change in concentration of metabolites after sample storage suggests that the biotransformation of parent drugs to metabolites still occurs during storage at positive temperatures. Quantitative results from long-term-stored urine samples with opioids or benzodiazepines should be interpreted cautiously. Ongoing monitoring of these patients will help the laboratory and clinician draw conclusions on patient adherence to their prescription regimen. In efforts to enable accurate interpretation of the toxicological results, it is important to know the stability of the drugs at several storage conditions and in different matrices. This study demonstrates the need for more detailed studies of in vitro changes of analytes in biological samples and in urine samples particularly. By considering other work related to this topic, and by conducting more research aimed at particular storage conditions, degradation models may be applied.
The purpose of this study was to investigate the stability of opioid and benzodiazepine drugs and metabolites in stored urine specimens. This is an area of interest for clinical and forensic toxicology applications. The data was collected under validated methodologies with quality control samples run with each batch. The results demonstrated that sample collection and storage conditions are important considerations in the context of drug testing.
- Allen KR (2011) Screening for drugs of abuse: which matrix, oral fluid or urine? Ann Clin Biochem 48:531–541View ArticleGoogle Scholar
- Anagnostis EA, Sadaka RE, Sailor LA, Moody DE, Dysart KC, Kraft WK (2011) Formulation of buprenorphine for sublingual use in neonates. J Pediatr Pharmacol Ther 16:281–284Google Scholar
- Anizan S, Bichon E, Duval T, Monteau F, Cesbron N, Antignac JP, Le Bizec B (2012) Gas chromatography coupled to mass spectrometry-based metabolomic to screen for anabolic practices in cattle: identification of 5alpha-androst-2-en-17-one as new biomarker of 4-androstenedione misuse. J Mass Spectrom 47:131–140View ArticleGoogle Scholar
- Atanasov VN, Stoykova S, Runiov A, Dimitrova T, Aleksandrova D, Tsakovski S, Mitewa M (2012) Stability of diazepam in blood samples at different storage conditions and in the presence of alcohol. Forensic Sci Int 215:159–163View ArticleGoogle Scholar
- Centers for Disease Control and Prevention (2012) Prescription painkiller overdoses in the US. VitalSigns, NovemberGoogle Scholar
- Cheng JY, Mok VK (2004) Rapid determination of ketamine in urine by liquid chromatography-tandem mass spectrometry for a high throughput laboratory. Forensic Sci Int 142:9–15View ArticleGoogle Scholar
- Cheng WC, Yau TS, Wong MK, Chan LP, Mok VK (2006) A high-throughput urinalysis of abused drugs based on a SPE-LC-MS/MS method coupled with an in-house developed post-analysis data treatment system. Forensic Sci Int 162:95–107View ArticleGoogle Scholar
- Clauwaert KM, Van Bocxlaer JF, De Leenheer AP (2001) Stability study of the designer drugs “MDA, MDMA and MDEA” in water, serum, whole blood, and urine under various storage temperatures. Forensic Sci Int 124:36–42View ArticleGoogle Scholar
- Cone EJ, Heit HA, Caplan YH, Gourlay D (2006) Evidence of morphine metabolism to hydromorphone in pain patients chronically treated with morphine. J Anal Toxicol 30:1–5View ArticleGoogle Scholar
- Couto JE, Romney MC, Leider HL, Sharma S, Goldfarb NI (2009) High rates of inappropriate drug use in the chronic pain population. Popul Health Manag 12:185–190View ArticleGoogle Scholar
- de Castro A, Concheiro M, Shakleya DM, Huestis MA (2009a) Development and validation of a liquid chromatography mass spectrometry assay for the simultaneous quantification of methadone, cocaine, opiates and metabolites in human umbilical cord. J Chromatogr B Analyt Technol Biomed Life Sci 877:3065–3071View ArticleGoogle Scholar
- de Castro A, Concheiro M, Shakleya DM, Huestis MA (2009b) Simultaneous quantification of methadone, cocaine, opiates, and metabolites in human placenta by liquid chromatography-mass spectrometry. J Anal Toxicol 33:243–252View ArticleGoogle Scholar
- de Jager AD, Bailey NL (2011) Online extraction LC-MS/MS method for the simultaneous quantitative confirmation of urine drugs of abuse and metabolites: amphetamines, opiates, cocaine, cannabis, benzodiazepines and methadone. J Chromatogr B Analyt Technol Biomed Life Sci 879:2642–2652View ArticleGoogle Scholar
- Drummer OH (2004) Postmortem toxicology of drugs of abuse. Forensic Sci Int 142:101–113View ArticleGoogle Scholar
- Drummer OH (2007) Post-mortem toxicology. Forensic Sci Int 165:199–203View ArticleGoogle Scholar
- El Mahjoub A, Staub C (2000) Stability of benzodiazepines in whole blood samples stored at varying temperatures. J Pharm Biomed Anal 23:1057–1063View ArticleGoogle Scholar
- Gasche Y, Daali Y, Fathi M, Chiappe A, Cottini S, Dayer P, Desmeules J (2004) Codeine intoxication associated with ultrarapid CYP2D6 metabolism. N Engl J Med 351:2827–2831View ArticleGoogle Scholar
- Hadidi KA, Oliver JS (1998) Stability of morphine and buprenorphine in whole blood. Int J Legal Med 111:165–167View ArticleGoogle Scholar
- Hammett-Stabler C, Cotten SW (2012) An introduction to drug testing: the expanding role of mass spectrometry. Methods Mol Biol 902:1–13View ArticleGoogle Scholar
- Hughes MM, Atayee RS, Best BM, Pesce AJ (2012) Observations on the metabolism of morphine to hydromorphone in pain patients. J Anal Toxicol 36:250–256View ArticleGoogle Scholar
- Kilpatrick GJ, Smith TW (2005) Morphine-6-glucuronide: actions and mechanisms. Med Res Rev 25:521–544View ArticleGoogle Scholar
- Klingmann A, Skopp G, Aderjan R (2001) Analysis of cocaine, benzoylecgonine, ecogonine methyl ester, and ecgonine by high-pressure liquid chromatography-API mass spectrometry and application to a short-term degradation study of cocaine in plasma. J Anal Toxicol 25:425–430View ArticleGoogle Scholar
- Lader MH (1999) Limitations on the use of benzodiazepines in anxiety and insomnia: are they justified? Eur Neuropsychopharmacol 9:S399–S405View ArticleGoogle Scholar
- Levine B, Blanke RV, Valentour JC (1983) Postmortem stability of benzodiazepines in blood and tissues. J Forensic Sci 28:102–115Google Scholar
- Liu Y, Uboh CE, Soma LR, Li X, Guan F, You Y, Chen JW (2011) Efficient use of retention time for the analysis of 302 drugs in equine plasma by liquid chromatography-MS/MS with scheduled multiple reaction monitoring and instant library searching for doping control. Anal Chem 83:6834–6841View ArticleGoogle Scholar
- Lu W, Bennett BD, Rabinowitz JD (2008) Analytical strategies for LC–MS-based targeted metabolomics. J Chromatogr B 871:236–242View ArticleGoogle Scholar
- Melo P, Bastos ML, Teixeira HM (2012) Benzodiazepine stability in postmortem samples stored at different temperatures. J Anal Toxicol 36:52–60View ArticleGoogle Scholar
- Moore KA, Ramcharitar V, Levine B, Fowler D (2003) Tentative identification of novel oxycodone metabolites in human urine. J Anal Toxicol 27:346–352View ArticleGoogle Scholar
- Mueller CA, Weinmann W, Dresen S, Schreiber A, Gergov M (2005) Development of a multi-target screening analysis for 301 drugs using a QTrap liquid chromatography/tandem mass spectrometry system and automated library searching. Rapid Commun Mass Spectrom 19:1332–1338View ArticleGoogle Scholar
- Peters FT (2007) Stability of analytes in biosamples - an important issue in clinical and forensic toxicology? Anal Bioanal Chem 388:1505–1519View ArticleGoogle Scholar
- Riss J, Cloyd J, Gates J, Collins S (2008) Benzodiazepines in epilepsy: pharmacology and pharmacokinetics. Acta Neurol Scand 118:69–86View ArticleGoogle Scholar
- Selden T, Roman M, Druid H, Kronstrand R (2011) LC-MS-MS analysis of buprenorphine and norbuprenorphine in whole blood from suspected drug users. Forensic Sci Int 209:113–119View ArticleGoogle Scholar
- Skopp G (2004) Preanalytic aspects in postmortem toxicology. Forensic Sci Int 142:75–100View ArticleGoogle Scholar
- Skopp G, Klingmann A, Potsch L, Mattern R (2001a) In vitro stability of cocaine in whole blood and plasma including ecgonine as a target analyte. Ther Drug Monit 23:174–181View ArticleGoogle Scholar
- Skopp G, Potsch L, Klingmann A, Mattern R (2001b) Stability of morphine, morphine-3-glucuronide, and morphine-6-glucuronide in fresh blood and plasma and postmortem blood samples. J Anal Toxicol 25:2–7View ArticleGoogle Scholar
- Smith RB, Kroboth PD, Vanderlugt JT, Phillips JP, Juhl RP (1984) Pharmacokinetics and pharmacodynamics of alprazolam after oral and IV administration. Psychopharmacology (Berl) 84:452–456View ArticleGoogle Scholar
- Vozeh S (1981) Pharmacokinetic of benzodiazepines in old age. Schweiz Med Wochenschr 111:1789–1793Google Scholar
- Yao X, Bajrami B, Shi Y (2010) Ultrathroughput multiple reaction monitoring mass spectrometry. Anal Chem 82:794–797View ArticleGoogle Scholar
- Yee DA, Best BM, Atayee RS, Pesce AJ (2012) Observations on the urine metabolic ratio of oxymorphone to oxycodone in pain patients. J Anal Toxicol 36:232–238View ArticleGoogle Scholar
- Yost RA, Enke CG (1979) Triple quadrupole mass spectrometry for direct mixture analysis and structure elucidation. Anal Chem 51:1251–1264View ArticleGoogle Scholar
- Zhou B, Xiao JF, Tuli L, Ressom HW (2012) LC-MS-based metabolomics. Mol BioSyst 8:470–481View ArticleGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.