- Open Access
ICH guideline practice: application of novel RP-HPLC-DAD method for determination of olopatadine hydrochloride in pharmaceutical products
© Basniwal and Jain; licensee Springer. 2013
- Received: 16 July 2013
- Accepted: 22 August 2013
- Published: 3 September 2013
A novel reverse-phase high-performance liquid chromatography (RP-HPLC)-DAD method was developed, validated and applied to quantify olopatadine hydrochloride in pharmaceutical products.
The RP-HPLC analyses were carried out using a mixture of 0.1% formic acid and methanol (35:65) as a mobile phase on ZORBAX Eclipse Plus C18 (250 mm × 4.6 mm, 5 μm), with a flow rate of 1.0 mL/min and UV detection at 300 nm. The developed method was validated as per the International Conference on Harmonisation of Technical Requirements for Registration of Pharmaceutical for Human Use guidelines for linearity, range, accuracy, precision, robustness, limit of detection, limit of quantitation and specificity.
The validated method was applied to quantify the drug content in raw material, tablets and eye drops which was found to be 99.36% to 101.02%, 98.99% to 100.56% and 99.09% to 100.91%, respectively.
The method was found to be efficient, precise, accurate, specific and economic and useful for the routine analysis of olopatadine hydrochloride in pharmaceutical industries.
- Olopatadine hydrochloride
- Pharmaceutical products
However, it had several problems, such as cross-reactivity of metabolites and low precision, and so it was considered that plasma concentrations of olopatadine measured by RIA needed to be confirmed by an alternative method. High-performance liquid chromatography (HPLC) with tandem mass spectrometry has been applied for the determination of a number of compounds in biological fluids (Zhu et al. 2011). Other assay methods for olopatadine and its metabolites in human plasma by HPLC with electrospray ionisation tandem mass spectrometry were reported (Fujimaki et al. 2006; Fujita et al. 1999). Varghese et al. (2011) have used 0.1% orthophosphoric acid (pH 4.5) with triethylamine/acetonitrile (75:25) for the determination of olopatadine hydrochloride. Triethylamine should be avoided as its addition to the mobile phase requires longer column equilibration times, which is not advisable for the routine use of the developed method. It occasionally introduces additional problems such as erratic baselines and poor peak shape and complicates the preparation of the mobile phase (Synder et al. 1997). A complicated buffer is a component of the mobile phase for the HPLC method, which includes monobasic potassium phosphate (6.8 g), 1-pentane sulphonic acid sodium salt monohydrate (1.28 g) and triethylamine (3 mL), and the pH is adjusted to 3.0 with orthophosphoric acid (Rele and Warkar 2011). Thus, both the above methods cannot be useful for routine analysis as tedious mobile phase composition and problem arise from triethylamine.
Hitherto, the objective of the present research work was to develop a simple, precise, accurate and fast reverse-phase high-performance liquid chromatography (RP-HPLC) assay method for the analysis of olopatadine hydrochloride in different pharmaceutical products including raw material, tablets and eye drops, and to compare the assay results by statistical analysis, which may be useful for routine analysis.
Instrumentation and chromatograph
The HPLC chromatograph used was Agilent Infinity 1260 series (Agilent Technologies, Inc., Santa Clara, CA, USA) equipped with 1260 binary pump VL (400 bar), 1260 manual injector (600 bar), Rheodyne 7725i seven-port sample injection valve with 20-μL fixed loop, ZORBAX Eclipse Plus C18 (250 mm × 4.6 mm, 5 μm), 1260 DAD VL, 20-Hz detector, standard flow cell (10 mm, 13 μL, 120 bar), OpenLab CDS EZChrom Ed. Workstation and syringe (50.0 μL, FN, LC tip). All weighing for analysis was performed on Shimadzu electronic analytical balance AX-200 (Kyoto, Japan). Water used for analysis was prepared by triple distillation assembly. All dilutions, mobile phase and other solutions that were used for the analysis were filtered through a 0.2-μm nylon filter (Ultipor®N66 Nylon 6,6 membrane, Pall Sciences, Pall India Pvt. Ltd., Mumbai, India).
Chemicals and reagent
The working standard used was OLO which was supplied by Ranbaxy Laboratories Limited (Gurgaon, Harayana, India) as a gift sample. Formic acid and methanol were procured from Merck Specialties Private Limited (Mumbai, India). The mobile phase was prepared from the combination of formic acid (0.1%) and methanol.
Accurately weighed 50 mg of OLO was dissolved in 50% aqueous methanol to prepare stock I (1,000 μg/mL) in a 50-mL volumetric flask. Stock II (100 μg/mL) was prepared from stock I, which was used to prepare further dilutions containing 1, 5, 10, 15 and 20 μg/mL. All dilutions were filtered through the 0.2-μm nylon filter (Ultipor®N66 Nylon 6,6 membrane, Pall Sciences) and chromatographed by a set of conditions on Agilent Infinity 1260 series. The mixture of 0.1% formic acid and methanol (35:65) was used as the mobile phase for the elution of the drug on Zorbax Eclipse Plus C18 column (250 mm × 4.6 mm, 5 μm) at 1.0 mL/min flow rate. OLO was successfully eluted at 3.77 min with a run time of 7 min, and detection was performed using a photodiode array detector (PDA) at 300 nm.
According to the International Conference on Harmonisation of Technical Requirements for Registration of Pharmaceutical for Human Use (ICH) guidelines (ICH 2000a, b), the developed method was validated to assure the reliability of the results of analysis for the different parameters, viz. linearity, range, accuracy, precision, robustness, limit of quantization (LOQ), limit of detection (LOD) and specificity. Linearity was determined by serial dilutions (1 to 20 μg/mL) of the OLO in 50% aqueous methanol in triplicate. The range of OLO was validated between 5 and 15 μg/mL with triplicates of dilutions. The recovery method was used to determine the accuracy of the developed method by spiking the standard solution to the pre-analysed samples (5, 10 and 15 μg/mL), which was repeated six times. Repeatability and intermediate precision were studied to assure the precision of the method. Six replicates of the standard dilution (5 μg/mL) were chromatographed subsequently to assure repeatability, which were further chromatographed after 5 h to assure the stability of the drug in the solvent system. The standard dilutions were analysed by different analysts in subsequent days to study intermediate precision in the linearity range. A change in temperature (20°C, 25°C and 30°C) and acidic content of the aqueous part of the mobile phase (5% change in 0.1% aqueous formic acid) was observed. Different serial dilutions of OLO (0.1 to 1,000 ng/mL) were chromatographed to calculate the signal-to-noise ratio to determine LOD and LOQ. Alkaline-degraded OLO samples were chromatographed to ascertain the specificity of the developed method for OLO.
Analysis of pharmaceutical products
As per the ‘Chromatography’ subsection, the samples of raw material have been prepared and analysed in the set of condition of analysis, which was repeated in six batches. The response to chromatographic analysis was used to calculate the concentration with the help of a regression equation.
Powdered tablets were weighed equivalently to 50 mg of OLO and sonicated to dissolve the drug content in methanol to extract the complete drug content from the tablet powder. The sonicated solution was filtered through Whatman filter no. 41 to remove the un-dissolved excipients of the tablet dosage form, and different serial dilutions were prepared by subsequent dilution with 50% aqueous methanol. All dilutions were filtered through the 0.2 μm nylon filter and chromatographed. The drug content was determined from the regression equation.
An accurately measured volume of ophthalmic solution (Olodin, FDC Limited, Aurangabad, India), equivalent to about 10 mg of OLO, was transferred to a 100 mL 163 volumetric flask and diluted with diluent to the volume. The solution was mixed and sonicated for 10 min. An appropriate dilution was prepared from the stock solution and filtered (stock P, 100 μg/mL). Aliquots of stock P were diluted to get sample concentrations (5, 10 and 15 μg/mL) in the range of linearity for the developed method and filtered through the 0.2 μm nylon filter. All these dilutions were chromatographed, and the area under the curve (AUC) of the peak of OLO was placed in the regression equation to get the concentration of the samples.
Optimization of chromatography
The aim of this research work was to develop a novel RP-HPLC method for the determination of OLO in different matrices by eliminating the triethylamine component in the mobile phase, which improves the peak shape associated with other number of troubles to the column. Keeping the view of the results with triethylamine use in the mobile phase, the physical properties of OLO were used to improve the chromatographic parameters. On the basis of the solubility of OLO, different combinations of the mobile phase were tried to elute on C18 column. Broadening and splitting of the peak have been seen with the mixture of methanol and water (50:50) and the mixture of acetonitrile and water (50:50). The pK a value (4.29) has been taken into account to design the mobile phase composition as OLO has a carboxylic group. So 0.1% formic acid was added to the aqueous phase of the mixture of acetonitrile and water (50:50), but the same broadening of peak with a little bit splitting of peak was observed. Tailing was found with the mixture of methanol and water (50:50) along with the acidic content of the aqueous phase which was improved by increasing the concentration of methanol content in the mobile phase. A very sharp peak shape was found with 70% content of methanol in the mobile phase, but it was merging with the diluent peak when a higher value was present at 60%. Thus, chromatographic conditions were concluded to be better than the reported methods.
The peak purity of OLO in the chromatogram was 1 unit which was supported by the peak purity curve (Figure 2D). A three-dimensional view of the chromatogram was the proof for no interference to the peak of OLO (Figure 2E).
System suitability parameters
System suitability parameters for OLO in 0.1% formic acid and methanol (35:65) on C18 column
Value (n= 6)
Number of theoretical plates
A linear relationship was found between the OLO concentrations and the response to chromatographic analysis which was demonstrated by high regression coefficients (r2 = 0.999) for the concentration of 1 to 20 μg/mL and the mean regression equation was found as AUC = 79,991X − 64,340. The working range for the drug was selected to be 5 to 15 μg/mL. Accuracy was investigated by the recovery of standard addition to pre-analysed samples which was near 100% with acceptable %RSD. Intra-day variation (repeatability) and intermediate precision (days and analyst variation) were evaluated by analysing quality control samples containing low, medium and high concentrations of OLO.
For intra-day variation, sets of the six replicates of the three concentrations were analysed on the same day; for intermediate precision, the six replicates were analysed on three different days by three different analysts. All results of precision were between 99.89% and 100.08% within the acceptable limits of %RSD values, and there was no significant difference in the results of the analysis when drug samples were analysed after 5 h. The developed method is robust when it was unaffected by small changes in operating conditions of the analysis. The experimental conditions were deliberately altered at three different levels, and chromatographic response was evaluated.
Validation parameters for olopatadine hydrochloride
Values a± SD, ± %RSD
1 to 20 μg/mL
AUC = 79,991X − 64,340
r2 = 0.999 ± 0.001, ±0.100
5 to 15 μg/mL ± 1.08, ±0.51
100.11 ± 1.065, ±0.784
99.94 ± 0.742, ±0.482
100.08 ± 0.825, ±0.091
Analyst to analyst
99.89 ± 0.929, ±0.027
Temperature (20°C, 25°C and 30°C)
99.91 ± 0.816, ±0.089
Formic acid concentration (±5%)
100.06 ± 0.794, ±0.296
5 ng/mL ± 1.098, ±0.566
0.5 ng/mL ± 1.691, ±0.295
Ascertained by analysing standard drug and degraded sample of equivalent concentration
Analysis of products
The validated chromatographic method was applied to determine OLO in raw material, tablets and eye drops, where six batches were prepared for each quality control sample (5, 10, and 15 μg/mL) for all pharmaceutical products.
Analysis of olopatadine hydrochloride in pharmaceutical products
Concentration of quality control samples (%)
Raw material (μg/mL)
Eye drop (μg/mL)
Hence, the simple, fast, precise, accurate and novel reverse-phase liquid chromatographic method was developed and validated according to ICH guidelines. The validated method was successfully applied to determine the OLO content in pharmaceutical products including raw material, tablets and eye drops with acceptable limits. As previous reported methods may not be applied for routine analysis due to the use of the triethylamine component in the mobile phase, this novel RP-HPLC method might be applied for assays, dissolution studies, bio-equivalence studies as well as routine analysis of OLO in pharmaceutical industries.
One of the authors, Pawan Kumar Basniwal, is earnestly indebted to the Science and Engineering Research Board (SERB), DST, New Delhi, for the financial support for this research work under Fast Track Scheme for Young Scientists. The authors are highly thankful to the Head of the School of Pharmaceutical Sciences, RGPV, Bhopal, and the Principal of the LBS College of Pharmacy, Jaipur, for providing the experimental facilities for this research work.
- Fujimaki K, Lee XP, Kumazawa T, Sato J, Sato K: Determination of some anti-allergic drugs in human plasma by direct-injection high-performance liquid chromatography-tandem mass spectrometry. Forensic Toxicol 2006, 24: 8–16.View ArticleGoogle Scholar
- Fujita K, Magara H, Kobayashi H: Determination of olopatadine, a new anti-allergic agent, and its metabolites in human plasma by high-performance liquid chromatography with electrospray ionization tandem mass spectrometry. J Chromatogr B 1999, 731: 345–352. S0378–4347(99)00236–4View ArticleGoogle Scholar
- ICH: Q2A: text on validation of analytical procedures. Geneva: International Conference on Harmonisation of Technical Requirements for Registration of Pharmaceutical for Human Use; 2000a.Google Scholar
- ICH: Q2B: validation of analytical procedure: methodology. Geneva: International Conference on Harmonisation of Technical Requirements for Registration of Pharmaceutical for Human Use; 2000b.Google Scholar
- Kajita J, Inano K, Fuse E, Kuwabara T, Kobayashi H: Effects of olopatadine, a new anti-allergic agent, on human liver microsomal cytochrome p450 activities. Drug Metabo Disposit 2002, 30: 1504–1511.View ArticleGoogle Scholar
- McGill JI: A review of the use of olopatadine in allergic conjunctivitis. Int Ophthalmology 2004,25(3):171–179.View ArticleGoogle Scholar
- Ohishi T, Magara H, Yasuzawa T, Kabayashi H, Yamaguchi K, Kobayashi S: Disposition of KW-4679 (4): metabolism of KW-4679 in rats and dogs. Xenobiol Metab Dispos 1995, 10: 689–706.Google Scholar
- Ohshima E, Sato H, Obase H, Uchimura T, Kuwabara T, Kobayashi S: Synthesis of a dibenz [b, e]oxepin-bovine serum albumin conjugate for radioimmunoassay of KW-4679 ((Z)-11-[3-(dimethylamino)propylidene]-6, 11-dihydrodibenz[b, e]oxepin-2-acetic acid hydrochloride). Chem Pharm Bull 1992, 40: 2552–2554.View ArticleGoogle Scholar
- Rele RV, Warkar CB: Application of high performance liquid chromatographic technique for olopatadine hydrochloride and its impurity in ophthalmic solution. Int J Chem Sci 2011,9(2):601–614.Google Scholar
- Scoper SV, Berdy GJ, Lichtenstein SJ, Rubin JM, Bloomenstein M, Prouty RE, Vogelson CT, Edwards MR, Waycaster C, Pasquine T, Gross RD, Robertson SM: Perception and quality of life associated with the use of olopatadine 0.2% (Pataday™) in patients with active allergic conjunctivitis. Adv Ther 2007, 24: 1221–1232.View ArticleGoogle Scholar
- Synder LR, Kirkland JJ, Glajch JL: Practical HPLC method development. 2nd edition. New York: Wiley; 1997:430.View ArticleGoogle Scholar
- Takahashi H, Zhang Y, Morita E: Evaluation of the anti-histamine effects of olopatadine, cetirizine and fexofenadine during a 24 h period: a double-blind, randomized, crossover, placebo-controlled comparison in skin responses induced by histamine iontophoresis. Arch Dermatol Res 2008, 300: 291–295.View ArticleGoogle Scholar
- Tsunoo M, Momomura S, Masuo M, Iizuka H: Phase I clinical study on KW-4679, an anti-allergic drug: safety and pharmacokinetics in the single and repeated administration study to healthy subjects. Kiso To Rinsho 1995, 29: 4129–4147.Google Scholar
- Varghese SJ, Kumar AM, Ravi TK: Stability-indicating high-performance column liquid chromatography and high-performance thin-layer chromatography methods for the determination of olopatadine hydrochloride in tablet dosage form. J AOAC Int 2011,94(6):1815–1820.View ArticleGoogle Scholar
- Zhu P, Wen Y, Fan XP, Zhou ZL, Fan RX, Chen JM, Huang KL, Zhu XL, Chen YF, Zhuang J: A rapid and sensitive liquid chromatography–tandem mass spectrometry method for determination of olopatadine concentration in human plasma. J Ana Toxicol 2011, 35: 113–118.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.