HPLC-CUPRAC post-column derivatization method for the determination of antioxidants: a performance comparison between porous silica and core-shell column packing
© The Author(s). 2018
Received: 11 August 2017
Accepted: 9 January 2018
Published: 17 January 2018
An HPLC method employing a post-column derivatization strategy using the cupric reducing antioxidant capacity reagent (CUPRAC reagent) for the determining antioxidants in plant-based materials leverages the separation capability of regular HPLC approaches while allowing for detection specificity for antioxidants.
Three different column types, namely core-shell and porous silica including two chemically different core-shell materials (namely phenyl-hexyl and C18), were evaluated to assess potential improvements that could be attained by changing from a porous silica matrix to a core-shell matrix. Tea extracts were used as sample matrices for the evaluation specifically looking at catechin and epigallocatechin gallate (EGCG).
Both the C18 and phenyl-hexyl core-shell columns showed better performance compared to the C18 porous silica one in terms of separation, peak shape, and retention time. Among the two core-shell materials, the phenyl-hexyl column showed better resolving power compared to the C18 column.
The CUPRAC post-column derivatization method can be improved using core-shell columns and suitable for quantifying antioxidants, exemplified by catechin and EGCG, in tea samples.
The importance of functional nutrition has sparked an active interest in natural antioxidants due to their inherent ability to protect the human body by inhibiting free radical reactions. As a corollary to this, the general public would want to know the constituents and levels of antioxidant capacity in the food we consume. There are numerous published methods claiming to measure total antioxidant capacity (TAC) in vitro; from the list of TAC assays in the literature, the ferric-reducing antioxidant power (FRAP) and cupric reducing antioxidant out (CUPRAC) methods are two most commonly used and are proposed for standardization during the First International Congress on Antioxidant Methods. The reaction mechanism in both the FRAP and CUPRAC methods toward antioxidants is solely electron transfer, and therefore, both methods can provide insight into dominant reaction mechanisms of various antioxidants in a given sample matrix, thus providing a framework for reliable quantification of total antioxidant activity (Prior et al. 2005).
Between the FRAP and CUPRAC methods, the latter has gained significant attention in the past few years because of its advantages in terms of its applicability for both hydrophilic and lipophilic antioxidants, its workability at physiological pH and its faster response (Huang et al. 2005). Since its development by Apak et al., the CUPRAC method for determining antioxidant capacity was employed numerous time by the different groups on various natural products of fruits and vegetables. For instance, TAC levels in multiple samples of apple juice were compared to individual antioxidant levels separately analyzed by HPLC (Karaman et al. 2010). The same approach was also employed to correlate the TAC levels with high-performance liquid chromatography (HPLC) findings for antioxidants present in the leaves of Petroselinum sativum (parsley), leaves of Apium graveolens (celery), and in Urtica dioica (stinging nettle) (Yıldız et al. 2008). In addition, the CUPRAC method for TAC determination has also been employed for the determination of trace level endogenous antioxidants in human serum with proven superior performance than FRAP especially for thiol-containing antioxidants like reduced glutathione (Apak et al. 2010). With its proven performance as a “total assay,” the CUPRAC reagent is an attractive candidate as a derivatization reagent in an HPLC post-column instrumental setup. Conceptually, the separated antioxidant components are allowed to react with the CUPRAC reagent in a post-column mixing unit before reaching the detector, thus leveraging the separation capability of regular HPLC approaches while allowing for detection specificity for antioxidants (Zacharis and Tzanavaras 2013). Multiple variants of this general post-column derivatization approach has been reported in the analysis of antioxidants from the leaves of Eucommia ulmoides Oliv. (Dai et al. 2013), from the flower buds of Lonicer spp. (Li et al. 2012) and from the leaves of Sonchus oleraceus Linn (Ou et al. 2013). All three reports used DPPH as the post-column derivatization reagent. Recently Jones and co-workers reported the use of FRAP reagent for the analysis of antioxidants in coffee (Jones et al. 2017). While proven to be functional, the main issue with HPLC post-column derivatization approaches, in general, is that the more post-column volume the HPLC system has (a necessity for analytes to react with the derivatization reagent), the broader the peaks will be which results in reduced chromatographic resolution.
We report here a method employing a similar post-column approach using CUPRAC as the post-column derivatization reagent. Although the general approach of this technique is not particularly new, improvements could potentially be attained if the column used was changed from a porous silica matrix to a core-shell matrix. The particle morphology of a core-shell column matrix is touted to result in less band broadening compared to fully porous particles and thus presumably delivers extremely high efficiencies and greater resolution (Gritti and Guiochon 2012b). This is a fundamentally logical assumption because in order to maximize efficiency, sources of band broadening need to be minimized. With core-shell particles, sources of band broadening (specifically the A and B terms) as described by the van Deemter equation are reduced compared to fully porous particles. This reduction in band broadening fundamentally results in chromatographic separations with better resolution, higher sensitivity, and improved peak capacities (Gritti and Guiochon 2012a). To our knowledge, this is the first report comparing the porous silica matrix with the core-shell matrix in a post-column derivatization format.
All the chemicals used in the experiments were purchased from either Sigma-Aldrich or TCI America and used as supplied. The HPLC grade solvents were purchased from VWR analytical, USA. The water used for the experiments was purified with a Milli-Q water system (Millipore, USA). Copper (II) chloride dihydrate, neocuproine hemihydrate, and sodium acetate used in the experiments were of the highest purity (> 99%) available. The antioxidants (+) catechin hydrate and (−) epigallocatechin gallate hydrate standards were used as standards for calibration and spike experiments.
Preparation of CUPRAC reagent
The CUPRAC reagent was prepared by mixing equal volume of 1 mM CuCl2*2H2O (in methanol/water 50/50), 25 mM Sodium acetate (in acetone/water 25/75), and 2 mM neocuproine (in methanol/water 50/50). The resultant reagent which itself is pale yellow in color, upon reacting with antioxidants, turns bright orange showing a maximum at 450 nm in UV. The reagent was prepared fresh prior to any experiment.
The UV experiments were done in a UV Microplate Reader (Spectra MAX 140 series, Molecular Devices). Spectra of catechin and EGCG (0.75 mM each in CH3OH/H2O 1:1) were taken with and without presence of CUPRAC reagent. Aliquots of 150 μL of each antioxidant (1.5 mM) were taken into a 96-well UV plate in different time intervals and mixed with 150 μL of CUPRAC reagent, and reading was taken immediately after shaking on an Eppendorf Mixmate shaker for 45 s. A control measurement (for dilution effect) was also performed in which, instead of the antioxidant solution, 150 μL of methanol/water 50/50 was mixed with the CUPRAC reagent and used as plate blank.
The antioxidants were analyzed using an HPLC instrument (Shimadzu LC 2040C) with a post-column mixing pump and a photodiode array (PDA) detector. The chromatographic separation was carried out in a porous silica column (Phenomenex Luna C18, 150 × 4.6 mm, 5 μm, Phenomenex, California, USA) and two core-shell columns (Phenomenex Kinetex C18, 150 × 4.6 mm, 5 μm and Phenomenex Kinetex phenyl-hexyl, 150 × 4.6 mm, 5 μm, Phenomenex, California, USA). Chromatographic separation was carried out using the following gradient system: solvent A, DI water, solvent B, and methanol. Gradient elution profile: 9-min linear gradient to 22% solvent B- 78% solvent A; isocratic elution for 6 min; gradient elution to 30% solvent B- 70% solvent A for another 5 min and return to initial condition via linear gradient for 3-min period. Flow rate of the separation gradient was 1.0 mL/min while that of the derivatization pump was 0.5 mL/min. The PDA detector was set at 450 nm for CUPRAC derivatization. The CUPRAC reagent used as post-column derivatization in the HPLC system is the same as previously described.
Preparation of tea samples
Tea samples of five commercial brands were procured from a local grocery store; brand 1 is a pure green tea, brand 2 is a blended black tea, brand 3 is a blended green tea, brand 4 is another blend of black tea, and brand 5 is chai black tea. Two grams of each brand were transferred into individual 150 mL Erlenmeyer flasks and extracted with 100 mL of boiled DI water. The mixtures were swirled occasionally, and the extraction was allowed to happen for 20 min. After that time, a 5-mL aliquot of each sample was transferred into a centrifuge tube and centrifuged at 4000 rpm for 10 min. The supernatant layer was syringe filtered using a 0.2-μm PTFE membrane before performing the experiments. Triplicate sample preparations were done on each sample, and two separate HPLC injections were carried out for each replicate preparation.
Results and discussion
Detection response–UV experiments
UV-vis spectroscopy was used in order to assess the spectral profiles of the samples and antioxidant standards and to reproduce spectral profiles reported in literature. The CUPRAC reagent is light turquoise with pale yellow overtone and does not show any observable peak between 350 and 650 nm. Upon introduction of an antioxidant, there was an observable change in color to yellow or deep orange in a concentration-dependent manner. It also exhibited concentration-dependent absorbance at 450 nm consistent with reports on CUPRAC assays (Özyürek et al. 2011; Apak et al. 2004). Similarly, when extracts from different tea samples–initially showing no observable peak in the visible region–were introduced into the CUPRAC reagent mixture, they showed peaks at 450 nm consistent with previous reports. Both experiments confirm the reduction of Cu(II)-neocuproine to yellow-colored Cu(I)-neocuproine complex consistent with observations reported in literature. Furthermore, the observed change in color–from turquoise to yellow/yellow orange–was almost instantaneous for both the standards catechin and EGCG as well as the tea extracts. This makes the CUPRAC reagent an attractive and compatible post-column derivatization reagent because the reaction kinetics of post-column derivatization reagents should ideally be fast (Zacharis and Tzanavaras 2013; Apak et al. 2005).
HPLC experiments: comparison of columns
Peak broadening due to “extra-column effects” has been recognized as one aspect in chromatographic separation that practitioners have thrived to minimize (Fekete and Fekete 2011; Gritti and Guiochon 2010). We have acquiesced to the inevitability of “extra-column effects”–and consequently to potential peak broadening due to extra-column volume––in this post-column derivatization format using CUPRAC, and hence, we looked at other potential avenues at which separation and quantification of antioxidants using this format can be improved. In this regard, columns packed with core-shell particles can be a good candidate.
Peak and separation characteristics for catechin and EGCG standards
Full width at half maximum (min)
Base width (W)
Retention time (t)
No. of theoretical plates (N)
C18 porous silica
0.470 ± 0.015
10.77 ± 0.01
2912 ± 170
C18 core shell
0.351 ± 0.057
8.73 ± 0.04
3431 ± 616
Phenyl-hexyl core shell
0.358 ± 0.054
8.59 ± 0.02
3193 ± 675
C18 porous silica
1.127 ± 0.052
15.02 ± 0.08
985 ± 71
C18 core shell
0.712 ± 0.121
12.47 ± 0.03
1701 ± 867
Phenyl-hexyl core shell
0.774 ± 0.089
14.19 ± 0.01
1864 ± 480
Although our present investigation is not exhaustive, a wealth of experimental data show that the measured column performance is strongly affected by extra-column band broadening especially pronounced in columns packed with particles smaller than the standard 4.6 μm (Fountain et al. 2009). Literature reporting sub-2 μm particle column characterization identified “extra-column effects” as a factor negative correlated with column performance (Heinisch et al. 2008; Usher et al. 2008; Cabooter et al. 2008).
Early reports on the benefits of using solid core materials in chromatographic columns have been associated with reduced transfer effects, the “C” term in the van Deemter equation. However, more recent reports indicate that these benefits were mostly contributed by dispersion processes, the “A” and “B” terms. The notion that solid core support materials provide a benefit to chromatographic performance by virtue of the A term (Eddy diffusion) is well supported with experimental data (Hayes et al. 2014). This is a very logical notion considering that the A term is directly related to particle size, structure, and packing efficiency of support particle of the column and solid core materials, pack better than fully porous materials. The solid core material provides a rougher surface and a better packing efficiency (Tanaka and McCalley 2016; Hayes et al. 2014). In our experiments, the particle sizes for both the porous silica and core shell are the same (5 μm) for the both column materials.
Furthermore, a significant reduction in dead volume is also expected in solid core materials compared to fully porous materials; a characteristic that relates to better column performance that can be theoretically explained with the B term. Gritti and co-workers reported that a fully porous material packed in a column will occupy only one-third of the column volume while the solid core material increases the occupied volume by 20–30% (Gritti et al. 2011). It is also recognized that the minimal value of longitudinal diffusion as reflected in the B term is obtained in part when the particles have negligible porosity or when the analyte is poorly retained or not retained at all. The limit of column improvement by virtue of the B term when using solid core materials is below the optimal linear velocity, i.e., low flow rates, as improvements are theoretically negligible at, or above, this optimal linear velocity (Hayes et al. 2014). The longitudinal diffusion (the B term) which contributes the most on band broadening at low velocity is associated with two parameters: the change in diffusion of the analyte in the presence of the stationary phase particles and the particle’s diffusion coefficient in the mobile phase. Gritti and co-workers simplified the B term based on the Garnett-Torquato model where it was shown that longitudinal diffusion can be minimized by increasing the ratio of solid core diameter to the whole diameter of the particle as it reduces the dead volume of the column (Gritti and Guiochon 2011; Gritti and Guiochon 2014). Although much of the benefits of the solid core derived from the A and B terms, the C term or resistance to mass transfer comes into effect for the bigger molecules and negligible for small ones (Gritti and Guiochon 2005; Gritti et al. 2006). Our experiments, by design, used EGCG (molar mass 458) being much bigger than catechin (molar mass 290) such that mass transfer effects could potentially be observed when separated in both fully porous and core-shell columns. However, these effects were not observed in our experiments. We presume that compared to the sizes of proteins and peptides which were intensively studied by Gritti and Guiochon, the sizes of the antioxidants used here may be too small to exhibit such mass transfer effects typically observed in proteins and peptides.
HPLC experiments: calibration and linearity
Given the differences in performance of the fully porous and core-shell columns, we proceeded to perform quantitative calibration to determine the linearity of concentration-dependent response and method detection limits for purposes of quantifying catechin and EGCG in tea samples. Calibration curves for the antioxidants catechin and EGCG were generated from fresh solutions and used for the quantification where each point in the calibration curve is the average of triplicate injection. The peak areas corresponding to each antioxidants were calculated automatically by the HPLC software (LabSolutions), and a linear relationship was established by plotting the average peak areas against concentration.
Linear regression, sensitivity, and limits of detection using different columns
Regression (r 2 )
C18 porous silica
6.30E + 05
C18 core shell
6.53E + 05
Phenyl-hexyl core shell
6.23E + 05
C18 porous silica
1.41E + 06
C18 core shell
1.42E + 06
Phenyl-hexyl core shell
1.49E + 06
At this point, it appears that the linear regression, sensitivity, and detection limits are comparable for all three columns when using this post-derivatization format. However, the fact that the peaks come out earlier for the core-shell columns with increased resolution relative to the porous silica column—despite the inevitable peak broadening due to “extra-column effects”—these characteristics of the core-shell columns could be leveraged to achieve higher analysis throughput without sacrificing sensitivity.
Quantification of antioxidants in tea
Antioxidants present per gram of tea samples (extracted in hot water for 20 min) separated in an HPLC with CUPRAC post-column derivatization using phenyl-hexyl core-shell column
Brand 1 (n = 3)
Brand 2 (n = 3)
Brand 3 (n = 3)
Brand 4 (n = 3)
Brand 5 (n = 3)
12.16 ± 0.24 (RSD = 1.97%)
0.68 ± 0.06 (RSD = 8.82%)
12.94 ± 0.17 (RSD = 1.31%)
0.45 ± 0.02 (RSD = 4.44%)
0.69 ± 0.05 (RSD = 7.25%)
40.71 ± 2.76 (RSD = 6.77%)
1.30 ± 0.01 (RSD = 0.78%)
61.57 ± 0.44 (RSD = 0.72%)
0.62 ± 0.16 (RSD = 25.81%)
1.09 ± 0.09 (RSD = 9.17%)
We have evaluated the performance of an HPLC method for the analysis of antioxidants with enhanced detection via post-column derivatization using CUPRAC employing three columns–two types of support matrices (porous silica and the core shell) and two types of stationary phases with the same support matrix (core shell, C18, and phenyl-hexyl stationary phases). Broad peaks were observed for all three columns which are a consequence of extra-column volume, but there was significantly less broadening in the core-shell column data. Theoretical considerations support the notion that the core-shell column would perform better than the porous silica column by virtue of the A and B terms in the van Deemter equation while the C term is negligible due to the small molecular sizes of the antioxidants. The C18 core-shell column exhibited sharper peaks with higher plate efficiency compared to the C18 porous silica column with shorter retention times. Among the two core-shell columns, the phenyl-hexyl column had better resolving power for the aromatic analytes which is predictable for its phenyl group. The separation and analysis of antioxidants in real samples–as demonstrated here using five brands of tea–can also benefit from the use of core-shell columns as it provides better and quicker separation (i.e., shorter retention times) compared to porous silica. While the linear regression, detection sensitivity, and detection limits are comparable for all three columns in this CUPRAC post-column derivatization approach, the characteristically shorter analyte retention times observed for the core-shell columns could be advantageous in achieving higher analysis throughput without sacrificing analytical sensitivity.
The authors gratefully acknowledge the USFDA-CTP for funding (RFA-FD-14-001). However, the views expressed in this paper do not necessarily reflect the official policies of the Department of Health and Human Services, nor do any mention of trade names, commercial practices, or organization imply endorsement by the US government.
This study was made possible, in part, through funds from the Food and Drug Administration through grant RFA-FD-14-001.
Availability of data and materials
Data, materials, and methods are all available for review and potential reproduction within the body of the paper under the “Methods” and “Results and discussion” sections. There is no separate supporting information provided for this paper.
SJPC, Ph.D. (Scientist) designed and conceptualized the study. SAH, Ph.D. (Postdoctoral scholar) conducted all experimental work and initial data interpretation. Both authors read and approved the final manuscript.
Ethics approval and consent to participate
We hereby declare that there are no human participation or human tissues used in this study, and ethics approval and consent to participate are thus waived.
Consent for publication
We hereby declare that no individual person’s data is gathered or used in this study in any form, and thus, consent for publication is hereby waived.
The authors declare that they have no conflict of interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Apak R, Güçlü K, Özyürek M, Bektaşoğlu B, Bener M. Cupric ion reducing antioxidant capacity assay for antioxidants in human serum and for hydroxyl radical scavengers. In: Armstrong D, editor. Advanced protocols in oxidative stress II. Totowa, NJ: Humana Press; 2010. p. 215–39.View ArticleGoogle Scholar
- Apak R, Güçlü K, Özyürek M, Karademir SE. Novel total antioxidant capacity index for dietary polyphenols and vitamins C and E, using their cupric ion reducing capability in the presence of neocuproine: CUPRAC method. J Agric Food Chem. 2004;52:7970–81.View ArticleGoogle Scholar
- Apak R, Güçlü K, Özyürek M, Karademir SE, Altun M. Total antioxidant capacity assay of human serum using copper(II)-neocuproine as chromogenic oxidant: the CUPRAC method. Free Radic Res. 2005;39:949–61.View ArticleGoogle Scholar
- Cabooter D, Billen J, Terryn H, Lynen F, Sandra P, Desmet G. Kinetic plot and particle size distribution analysis to discuss the performance limits of sub-2μm and supra-2μm particle columns. J Chromatogr A. 2008;1204:1–10.View ArticleGoogle Scholar
- Dai X, Huang Q, Zhou B, Gong Z, Liu Z, Shi S. Preparative isolation and purification of seven main antioxidants from Eucommia Ulmoides Oliv. (Du-zhong) leaves using HSCCC guided by DPPH-HPLC experiment. Food Chem. 2013;139:563–70.View ArticleGoogle Scholar
- Fekete S, Fekete J. The impact of extra-column band broadening on the chromatographic efficiency of 5cm long narrow-bore very efficient columns. J Chromatogr A. 2011;1218:5286–91.View ArticleGoogle Scholar
- Fountain KJ, Neue UD, Grumbach ES, Diehl DM. Effects of extra-column band spreading, liquid chromatography system operating pressure, and column temperature on the performance of sub-2-μm porous particles. J Chromatogr A. 2009;1216:5979–88.View ArticleGoogle Scholar
- Gritti F, Farkas T, Heng J, Guiochon G. On the relationship between band broadening and the particle-size distribution of the packing material in liquid chromatography: theory and practice. J Chromatogr A. 2011;1218:8209–21.View ArticleGoogle Scholar
- Gritti F, Felinger A, Guiochon G. Influence of the errors made in the measurement of the extra-column volume on the accuracies of estimates of the column efficiency and the mass transfer kinetics parameters. J Chromatogr A. 2006;1136:57–72.View ArticleGoogle Scholar
- Gritti F, Guiochon G. Critical contribution of nonlinear chromatography to the understanding of retention mechanism in reversed-phase liquid chromatography. J Chromatogr A. 2005;1099:1–42.View ArticleGoogle Scholar
- Gritti F, Guiochon G. On the extra-column band-broadening contributions of modern, very high pressure liquid chromatographs using 2.1mm I.D. columns packed with sub-2μm particles. J Chromatogr A. 2010;1217:7677–89.View ArticleGoogle Scholar
- Gritti F, Guiochon G. Diffusion models in chromatographic columns packed with fully and superficially porous particles. Chem Eng Sci. 2011;66:3773–81.View ArticleGoogle Scholar
- Gritti F, Guiochon G. Facts and legends about columns packed with sub-3-μm core-shell particles. LCGC N Am. 2012a;30:586–95.Google Scholar
- Gritti F, Guiochon G. Repeatability of the efficiency of columns packed with sub-3μm core–shell particles: part I. 2.6μm Kinetex-C18 particles in 4.6mm and 2.1mm×100mm column formats. J Chromatogr A. 2012b;1252:31–44.View ArticleGoogle Scholar
- Gritti F, Guiochon G. Rapid development of core–shell column technology: accurate measurements of the intrinsic column efficiency of narrow-bore columns packed with 4.6 down to 1.3μm superficially porous particles. J Chromatogr A. 2014;1333:60–9.View ArticleGoogle Scholar
- Hayes R, Ahmed A, Edge T, Zhang H. Core–shell particles: preparation, fundamentals and applications in high performance liquid chromatography. J Chromatogr A. 2014;1357:36–52.View ArticleGoogle Scholar
- Heinisch S, Desmet G, Clicq D, Rocca J-L. Kinetic plot equations for evaluating the real performance of the combined use of high temperature and ultra-high pressure in liquid chromatography. J Chromatogr A. 2008;1203:124–36.View ArticleGoogle Scholar
- Huang D, Ou B, Prior RL. The chemistry behind antioxidant capacity assays. J Agric Food Chem. 2005;53:1841–56.View ArticleGoogle Scholar
- Jones A, Pravadali-Cekic S, Dennis GR, Bashir R, Mahon PJ, Shalliker RA. Ferric reducing antioxidant potential (FRAP) of antioxidants using reaction flow chromatography. Anal Chim Acta. 2017;967:93–101.View ArticleGoogle Scholar
- Karaman Ş, Tütem E, Sözgen Başkan K, Apak R. Comparison of total antioxidant capacity and phenolic composition of some apple juices with combined HPLC–CUPRAC assay. Food Chem. 2010;120(4):1201–9.View ArticleGoogle Scholar
- Li Y-J, Chen J, Li Y, Li P. Identification and quantification of free radical scavengers in the flower buds of Lonicera species by online HPLC-DPPH assay coupled with electrospray ionization quadrupole time-of-flight tandem mass spectrometry. Biomed Chromatogr. 2012;26:449–57.View ArticleGoogle Scholar
- Ou Z-Q, Schmierer DM, Rades T, Larsen L, McDowell A. Application of an online post-column derivatization HPLC-DPPH assay to detect compounds responsible for antioxidant activity in Sonchus oleraceus L. leaf extracts. J Pharm Pharmacol. 2013;65:271–9.View ArticleGoogle Scholar
- Özyürek M, Güçlü K, Apak R. The main and modified CUPRAC methods of antioxidant measurement. TrAC Trends Anal Chem. 2011;30:652–64.View ArticleGoogle Scholar
- Prior RL, Wu X, Schaich K. Standardized methods for the determination of antioxidant capacity and phenolics in foods and dietary supplements. J Agric Food Chem. 2005;53:4290–302.View ArticleGoogle Scholar
- Tanaka N, McCalley DV. Core–shell, ultrasmall particles, monoliths, and other support materials in high-performance liquid chromatography. Anal Chem. 2016;88:279–98.View ArticleGoogle Scholar
- Usher KM, Simmons CR, Dorsey JG. Modeling chromatographic dispersion: a comparison of popular equations. J Chromatogr A. 2008;1200:122–8.View ArticleGoogle Scholar
- Yıldız L, Başkan KS, Tütem E, Apak R. Combined HPLC-CUPRAC (cupric ion reducing antioxidant capacity) assay of parsley, celery leaves, and nettle. Talanta. 2008;77(1):304–13.View ArticleGoogle Scholar
- Zacharis CK, Tzanavaras PD. Liquid chromatography coupled to on-line post column derivatization for the determination of organic compounds: a review on instrumentation and chemistries. Anal Chim Acta. 2013;798:1–24.View ArticleGoogle Scholar