- Technical note
- Open Access
Removal of traces of mercury from a carrier gas for analytical purpose
© Theil et al.; licensee Springer. 2013
Received: 5 July 2013
Accepted: 26 September 2013
Published: 11 November 2013
The analysis of mercury by cold vapor requires a gas, usually argon or helium, to transport elementary mercury to the gold trap or directly to the detector. When analyzing mercury in environmental matrices, a gas with a metal concentration as low as a few picograms per cubic meter is needed. Different sorbents have been used to purify the gas for a long time, but little information is available about them, mainly considering the analytical purpose. This paper presents results of the absorption capacity for solids and hypochlorite solutions that usually are used as mercury sorbents, giving technical information to the analyst to decide the best gas cleaning process to be used.
The absorption capacities of different sorbents were tested using atomic fluorescence spectrometry. Among the tested solids, platinum presented the highest absorption capacity (13.04 pg Hg per gram of Pt). Interaction between sodium hypochlorite, sodium chloride, and EDTA in the absorption capacity was investigated by a 23 factorial design. Results showed a significant interaction between hypochlorite and chloride.
A solution of 1.26 mmol L−1 sodium hypochlorite, 0.48 mol L−1 sodium chloride, and 0.6 mmol L−1 EDTA shows the highest absorption capacity (167.3 pg Hg) among the tested compositions. That solution has eliminated even traces of mercury from gases, resulting in a carrier free of mercury, what cannot be achieved using the solid sorbents tested, despite the use of solutions which is more tedious than the use of solids to clean gas. Anyway, the hypochlorite solution shows to be a good option to clean gases that have to be used in the analysis of mercury in samples with very low concentration.
The determination of the concentration of metals in environmental samples has drawn the attention of researchers over the years, and the chemical speciation analyses of metals have received particular attention (Borges et al. 2012; Wotter et al. 2011). The impact of mercury in the environment concerning its toxicity, mobility, bioaccumulation, and biomagnification is responsible for a continuous interest to investigate the fate of this metal (Santos et al. 2008; Stergarsek et al. 2013; Emili et al. 2012). The development of analytical methods and lower limits of quantification have allowed increasing the understanding of biogeochemical processes involving mercury and other metals. Among the most popular analytical methods to analyze mercury is the cold vapor atomic fluorescence spectrometry (CVAFS) (Santos et al. 2008; Leermakers et al. 2005), whose main advantage is separation of the analyte from the matrix by purging the volatile mercury with an inert gas to the detector, with a net result of high sensitivity and selectivity and lower limit of quantification. However, the success of the determination by CVAFS is closely related to the quality of the gas used to purge the analyte, since it is often necessary to make determinations of a few picograms of mercury (Liang et al. 1996).
Concentrations of dissolved gaseous mercury in surface water or dimethylmercury in environmental matrices, for example, are extremely low, and therefore, the need for extremely low blank values is essential (Emili et al. 2012; Hines et al. 2012). The care with cleaning materials, purification of reagents, and nature of the materials to be used in the mercury analyses are perfectly defined (MESL 1997). However, if the mercury concentration in the gas is not low enough, one should purify it.
The gas purification can be achieved by different ways. Solid sorbents, as traps containing particles of gold, platinum, or active carbon, usually are used by the analyst due to their operational facility (Liang et al. 1996). Solutions also can be used to remove mercury from gaseous streams. The mercury can be oxidized by hypochlorite solutions, producing Hg2+ (Sizeneva et al. 2005), but the importance of the physical states during the reaction must be remembered. Thus, the dissolution of one drop of metallic mercury in a hypochlorite solution is a heterogeneous reaction which takes place on the surface, depending on the area and the surface state. Sizeneva et al. (2005) established that the reaction of mercury dissolution in hypochlorite solution is first order. If mercury is considered to be in the gaseous state reacting with the hypochlorite solution, then the reaction is second order (Zhao and Rochelle 1999).
This paper presents a study comparing the capacity of well-known sorbents to remove gaseous mercury and the mercury concentration obtained after a helium stream passed through those sorbents.
Materials and methods
The tests were carried out with helium gas 5.0 in which the mercury concentration was determined by atomic fluorescence spectrometry after absorption on a gold-coated sand trap (MESL 1997; Horvat et al. 1993) to be equal to 6.35 ng m−3 (n = 5). The concentration of a commercial sodium hypochlorite (0.21 mol L−1) was determined by iodometric titration (Lagowski 1995). Sodium chloride and EDTA were pro-analysis (Merck, Whitehouse Station, NJ, USA). Ultrapure water (resistivity 18.2 MΩ cm, Milli-Q, Millipore Co., Billerica, MA, USA) was used to prepare all the solutions.
where W1 is the total mass of mercury that flows through the system during the experiment time, calculated by multiplying the time of the experiment, mercury concentration in the gas, and the gas flow rate; W2 is the mass of mercury not retained in EHg (W2) calculated using the peak height.
Four different solid sorbents are tested, namely (a) coal taken from a chemical cartridge mask of a multi-gas personal protection with a bed height of 65 mm, here referred to as C1; (b) coal taken from a chemical cartridge to be used against mercury vapors or chlorine gas from an individual protective mask with a bed height of 60 mm, here referred to as C2; (c) metallic gold particles with an average diameter of 0.1 mm and bed height of 15 mm, referred to as A1; and (d) platinum wire coil with a bed height of 10 mm, referred to as A2. The solids are packed into columns, with quartz wool used to seal the ends of the column.
When solutions are evaluated to sorb mercury, the system is assembled as shown in Figure 1, and a 150-mL glass flask is used, containing 50 mL of the solution to be tested, in EHg position. The gas flows through the system during a period (TPC) determining the pre-concentration of mercury. Then, the metal is desorbed by heating, and the fluorescence is measured. If the TPC is low and no peak appears, the procedure has to be repeated until a peak is recorded. The total time is recorded, i.e., the total volume of gas, and thus, the total mass of mercury (W1) that flows through the system during the experiment time can be calculated, as well as the mass of Hg that EHg does not retain during the experiment. In the following, AC is calculated.
A 23 factorial design is used to evaluate the influence of the composition of the absorbent solution on the absorption capacity. All possible combinations between three factors (N = 3) and two levels of concentration (k = 2) are evaluated, i.e., a full factorial design is run. Three replicates for each combination (n = 3) are run. The following factors and levels are evaluated: (a) chloride concentration (0.48 and 0.96 mol L−1), (b) hypochlorite concentration (0.63 and 1.26 mmol L−1), and (c) EDTA concentration (0.30 and 0.60 mmol L−1). The analysis of variance is used to identify the significance and the interaction between the factors for a significance level of 5%.
The mass of mercury is calculated comparing the height of the peak of the samples with the height of the peak from the standard. The peak of the standard is recorded after an injection of an aliquot of a gas mixture containing a mercury concentration is well established, as described elsewhere (Gardfeldt et al. 2002). A 50-μL syringe gas tight with removable needle (model 1705, Hamilton, Reno, NV, USA) is used to transfer an aliquot of 10 μL of gas mixture kept at 6.5°C in a thermostatic bath (model MQBTC99-20, Microquimica, Florianópolis, Brazil), corresponding to a mass of 40.4 pg Hg. The aliquot is injected into the system through the septum in the SI (Figure 1). The analyte is pre-concentrated on a gold-coated sand trap, desorbed by heating, and the analytical signal corresponding to 40.4 pg Hg is then recorded.
Results and discussion
Performance of sorbents for removing mercury from a gas mixture
A1 + A2
The mutual influence of the concentrations of hypochlorite and chloride on mercury removal is experimentally investigated by a 23 factorial design with the following factors: concentration of hypochlorite, chloride, and EDTA, investigated at two levels of concentration.
The results in Figure 2 are used to define the hypochlorite concentrations to be tested, being chosen as 0.63 and 1.26 mmol L−1. The highest hypochlorite concentration is chosen based on the maximum AC seen in Figure 2, and the lowest concentration is chosen as 50% of that value.
Zhao and Rochelle (1999) have demonstrated that, when using 1.0 mol L−1 NaCl in the presence of sodium hypochlorite, the concentration of mercury in the gas phase is stabilized for their experiments. However, when using 0.1 mol L−1 NaCl, those authors proved that the mercury concentration gradually increased in the gas phase (Zhao and Rochelle 1999). Thus, to perform the tests, 0.48 and 0.96 mol L−1 chloride concentrations are chosen.
It is assumed that the use of EDTA can increase AC because EDTA ensures the stabilization of Hg2+ formed by redox reaction, since the logKest HgEDTA ranges from 21.5 to 23.5. The selected concentrations are 0.30 and 0.60 mmol L−1 to perform the tests, ensuring large excess of this reagent with respect to the concentration of Hg2+.
Results of ANOVA
Causes of variation
OCl− × EDTA
OCl− × Cl−
EDTA × Cl−
OCl− × EDTA × Cl−
The use of a column containing a platinum coil proves to be more suitable for removing gaseous mercury, among the solid sorbents tested. The use of a solution of 1.26 mmol L−1 OCl−, 0.48 mol L−1 Cl−, and 0.6 mmol L−1 EDTA has an absorption capacity of 167 pg for gaseous Hg. This solution can purify gases that shall be used to analyze mercury in environmental samples, but the higher the mercury concentration in the gas, the more frequent the substitution of the solution. It also should be pointed out that it is a very easy way to remove the gaseous mercury and to get good blanks, even when a high-purity gas is not available.
- Borges AR, Niencheski LFH, Milani ICB, Milani MR: Optimisation and application of the voltammetric technique for speciation of chromium in the Patos Lagoon Estuary-Brazil. Envir Moni Assess 2012, 184: 5553–5562. doi:10.1007/s10661–011–2361–7 10.1007/s10661-011-2361-7View ArticleGoogle Scholar
- Emili A, Acquavita A, Koron N, Covelli S, Faganell J, Horvat M, Zizek S, Fajon V: Benthic flux measurements of Hg species in a northern Adriatic lagoon environment (Marano and Grado Lagoon, Italy). Est Coast Schelf Sci 2012, 113: 71–85. doi:10.1016/j.ecss.2012.05.018View ArticleGoogle Scholar
- Gardfeldt K, Horvat M, Sommar J, Kotnik J, Fajon V, Wangberg I, Lindqvist O: Comparison of procedures for measurements of dissolved gaseous mercury in seawater performed on a Mediterranean cruise. Anal Bioanal Chem 2002, 374: 1002–1008. doi:10.1007/s00216–002–1592–4 10.1007/s00216-002-1592-4View ArticleGoogle Scholar
- Hines ME, Poitras EM, Covelli S, Faganeli J, Emili A, Zizek S, Horvat M: Mercury methylation and demethylation in Hg-contaminated lagoon sediments (Marano and Grado Lagoon, Italy). Est Coast Schelf Sci 2012, 113: 85–95. doi:10.1016/j.ecss.2011.12.021View ArticleGoogle Scholar
- Horvat M, Lian L, Bloom NS: Comparison of distillation with other current isolation methods for the determination of methyl mercury compounds in low level environmental samples. Part II. Water Anal Chim Acta 1993, 282: 153–168. doi:0003–2670/93/$06.00 10.1016/0003-2670(93)80364-QView ArticleGoogle Scholar
- Lagowski JJ: Experiences in chemistry. New York: McGraw-Hill; 1995.Google Scholar
- Leermakers M, Baeyens W, Quevauviller P, Horvat M: Mercury in environmental samples: speciation, artifacts and validation. Trends Anal Chem 2005, 24: 383–393. doi:10.1016/j.trac.2004.01.001 10.1016/j.trac.2004.01.001View ArticleGoogle Scholar
- Liang L, Horvat M, Danilchik P: A novel analytical method for determination of picogram levels of total mercury in gasoline and other petroleum based products. Sci Total Environ 1996, 87: 57–64. doi:0048–9697(96)05129–7View ArticleGoogle Scholar
- Marine Environmental Studies Laboratory - MESL: International Atomic Energy Agency. Monaco: Standard Operating Procedures; 1997.Google Scholar
- Santos EJ, Hermann AB, Frescura VLA, Sturgeon RE, Curtius AJ: A novel approach to cold vapor generation for the determination of mercury in biological samples. J Braz Chem Soc 2008, 19: 929–934. 10.1590/S0103-50532008000500018View ArticleGoogle Scholar
- Sizeneva IP, Val'tsifer VA, Strel'nikov VN: A study of mercury dissolution in aqueous solutions of sodium hypochlorite. Russ J Appl Chem 2005, 78: 546–548. doi:1070–4272/05/7804–0546 10.1007/s11167-005-0337-5View ArticleGoogle Scholar
- Stergarsek A, Horvat M, Frkal P, Guevara SR, Kocjancic R: Removal of Hg0 in wet FGD by catalytic oxidation with air – a contribution to the development of a process chemical model. Fuel 2013, 107: 183–191. doi:10.1016/j.fuel.2012.08.001View ArticleGoogle Scholar
- Vieira S, Hoffmann R: Estatística Experimental. São Paulo: Atlas SA; 1989.Google Scholar
- Wotter SET, Niencheski LFH, Milani MR: Chemical speciation and dissolved iron in the pore water of Patos Lagoon sediments - Brazil. P Electrochim Acta 2011, 29: 155–163. doi:10.4152/pea.201103155 10.4152/pea.201103155View ArticleGoogle Scholar
- Zhao LL, Rochelle GT: Mercury absorption in aqueous hypochlorite. Chem Eng Sci 1999, 54: 655–662. doi:0009–2509(98)00263–2 10.1016/S0009-2509(98)00263-2View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.